<?xml version="1.0" encoding="utf-8"?>
<!DOCTYPE HWPML [
	<!ENTITY nbsp	"&#160;">
]>
<HWPML Version="2.1"><HEAD SecCnt="1"><DOCSUMMARY><TITLE>의료기기 허가·신고·심사 등에 관한 규정</TITLE>
<AUTHOR>법제처 법령정보센터</AUTHOR>
<DATE>2017.07.24</DATE>
</DOCSUMMARY>
<DOCSETTING><BEGINNUMBER Page="1" Footnote="1" Endnote="1" Picture="1" Table="1" Equation="1"/>
<CARETPOS List="0" Para="0" Pos="24"/>
</DOCSETTING>
<MAPPINGTABLE><BINDATALIST Count="9"><BINITEM Type="Embedding" BinData="1" Format="bmp"/>
<BINITEM Type="Embedding" BinData="2" Format="bmp"/>
<BINITEM Type="Embedding" BinData="3" Format="bmp"/>
<BINITEM Type="Embedding" BinData="4" Format="gif"/><BINITEM Type="Embedding" BinData="5" Format="gif"/><BINITEM Type="Embedding" BinData="6" Format="gif"/><BINITEM Type="Embedding" BinData="7" Format="gif"/><BINITEM Type="Embedding" BinData="8" Format="gif"/>
<BINITEM Type="Embedding" BinData="9" Format="png"/>
</BINDATALIST>
<FACENAMELIST><FONTFACE Lang="Hangul" Count="5"><FONT Id="0" Type="ttf" Name="굴림"><TYPEINFO FamilyType="2" SerifStyle="11" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="1" Type="ttf" Name="바탕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="2" Type="ttf" Name="HY견고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="3" Type="ttf" Name="HY신명조"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="4" Type="ttf" Name="HY중고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
</FONTFACE>
<FONTFACE Lang="Latin" Count="5"><FONT Id="0" Type="ttf" Name="굴림"><TYPEINFO FamilyType="2" SerifStyle="11" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="1" Type="ttf" Name="바탕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="2" Type="ttf" Name="HY견고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="3" Type="ttf" Name="HY신명조"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="4" Type="ttf" Name="HY중고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
</FONTFACE>
<FONTFACE Lang="Hanja" Count="4"><FONT Id="0" Type="ttf" Name="굴림"><TYPEINFO FamilyType="2" SerifStyle="11" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="1" Type="ttf" Name="바탕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="2" Type="ttf" Name="HY견고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="3" Type="ttf" Name="HY중고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
</FONTFACE>
<FONTFACE Lang="Japanese" Count="4"><FONT Id="0" Type="ttf" Name="굴림"><TYPEINFO FamilyType="2" SerifStyle="11" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="1" Type="ttf" Name="바탕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="2" Type="ttf" Name="HY견고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="3" Type="ttf" Name="HY중고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
</FONTFACE>
<FONTFACE Lang="Other" Count="5"><FONT Id="0" Type="ttf" Name="굴림"><TYPEINFO FamilyType="2" SerifStyle="11" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="1" Type="ttf" Name="바탕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="2" Type="ttf" Name="HY견고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="3" Type="ttf" Name="HY신명조"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="4" Type="ttf" Name="HY중고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
</FONTFACE>
<FONTFACE Lang="Symbol" Count="5"><FONT Id="0" Type="ttf" Name="굴림"><TYPEINFO FamilyType="2" SerifStyle="11" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="1" Type="ttf" Name="바탕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="2" Type="ttf" Name="HY견고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="3" Type="ttf" Name="HY신명조"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="4" Type="ttf" Name="HY중고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
</FONTFACE>
<FONTFACE Lang="User" Count="5"><FONT Id="0" Type="ttf" Name="굴림"><TYPEINFO FamilyType="2" SerifStyle="11" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="1" Type="ttf" Name="바탕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="2" Type="ttf" Name="HY견고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="3" Type="ttf" Name="HY신명조"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
<FONT Id="4" Type="ttf" Name="HY중고딕"><TYPEINFO FamilyType="2" SerifStyle="3" Weight="6" Proportion="0" Contrast="0" StrokeVariation="1" ArmStyle="1" Letterform="1" Midline="1" XHeight="1"/>
</FONT>
</FONTFACE>
</FACENAMELIST>
<BORDERFILLLIST Count="11"><BORDERFILL Id="1" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="None" Width="0.1mm" Color="0"/>
<RIGHTBORDER Type="None" Width="0.1mm" Color="0"/>
<TOPBORDER Type="None" Width="0.1mm" Color="0"/>
<BOTTOMBORDER Type="None" Width="0.1mm" Color="0"/>
<DIAGONAL Type="Solid" Width="0.1mm" Color="0"/>
<FILLBRUSH><WINDOWBRUSH FaceColor="4294967295" HatchColor="0"/>
</FILLBRUSH>
</BORDERFILL>
<BORDERFILL Id="2" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<RIGHTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<TOPBORDER Type="Solid" Width="0.12mm" Color="0"/>
<BOTTOMBORDER Type="Solid" Width="0.12mm" Color="0"/>
<DIAGONAL Type="Solid" Width="0.1mm" Color="0"/>
</BORDERFILL>
<BORDERFILL Id="3" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="None" Width="0.12mm" Color="0"/>
<RIGHTBORDER Type="None" Width="0.12mm" Color="0"/>
<TOPBORDER Type="Solid" Width="0.12mm" Color="0"/>
<BOTTOMBORDER Type="None" Width="0.12mm" Color="0"/>
<DIAGONAL Type="Solid" Width="0.1mm" Color="0"/>
</BORDERFILL>
<BORDERFILL Id="4" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="None" Width="0.12mm" Color="0"/>
<RIGHTBORDER Type="None" Width="0.12mm" Color="0"/>
<TOPBORDER Type="None" Width="0.12mm" Color="0"/>
<BOTTOMBORDER Type="Solid" Width="0.12mm" Color="0"/>
<DIAGONAL Type="Solid" Width="0.1mm" Color="0"/>
</BORDERFILL>
<BORDERFILL Id="5" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="None" Width="0.12mm" Color="0"/>
<RIGHTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<TOPBORDER Type="None" Width="0.12mm" Color="0"/>
<BOTTOMBORDER Type="None" Width="0.12mm" Color="0"/>
<DIAGONAL Type="Solid" Width="0.1mm" Color="0"/>
</BORDERFILL>
<BORDERFILL Id="6" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<RIGHTBORDER Type="None" Width="0.12mm" Color="0"/>
<TOPBORDER Type="None" Width="0.12mm" Color="0"/>
<BOTTOMBORDER Type="None" Width="0.12mm" Color="0"/>
<DIAGONAL Type="Solid" Width="0.1mm" Color="0"/>
</BORDERFILL>
<BORDERFILL Id="7" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<RIGHTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<TOPBORDER Type="None" Width="0.12mm" Color="0"/>
<BOTTOMBORDER Type="None" Width="0.12mm" Color="0"/>
<DIAGONAL Type="Solid" Width="0.1mm" Color="0"/>
</BORDERFILL>
<BORDERFILL Id="8" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<RIGHTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<TOPBORDER Type="None" Width="0.12mm" Color="0"/>
<BOTTOMBORDER Type="Solid" Width="0.12mm" Color="0"/>
<DIAGONAL Type="Solid" Width="0.1mm" Color="0"/>
</BORDERFILL>
<BORDERFILL Id="9" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<RIGHTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<TOPBORDER Type="Solid" Width="0.12mm" Color="0"/>
<BOTTOMBORDER Type="None" Width="0.12mm" Color="0"/>
<DIAGONAL Type="Solid" Width="0.1mm" Color="0"/>
</BORDERFILL>
<BORDERFILL Id="10" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<RIGHTBORDER Type="Solid" Width="0.12mm" Color="0"/>
<TOPBORDER Type="Solid" Width="0.12mm" Color="0"/>
<BOTTOMBORDER Type="Solid" Width="0.12mm" Color="0"/>
</BORDERFILL>
<BORDERFILL Id="11" ThreeD="false" Shadow="false" Slash="0" BackSlash="0" CrookedSlash="0"><LEFTBORDER Type="None" Width="0.12mm" Color="0"/>
<RIGHTBORDER Type="None" Width="0.12mm" Color="0"/>
<TOPBORDER Type="None" Width="0.12mm" Color="0"/>
<BOTTOMBORDER Type="None" Width="0.12mm" Color="0"/>
</BORDERFILL>
</BORDERFILLLIST>
<CHARSHAPELIST Count="26"><CHARSHAPE Id="0" Height="1000" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RATIO Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="1" Height="1000" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="1" Latin="1" Hanja="1" Japanese="1" Other="1" Symbol="1" User="1"/>
<RATIO Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="2" Height="900" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RATIO Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="3" Height="900" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="1" Latin="1" Hanja="1" Japanese="1" Other="1" Symbol="1" User="1"/>
<RATIO Hangul="95" Latin="95" Hanja="95" Japanese="95" Other="95" Symbol="95" User="95"/>
<CHARSPACING Hangul="-5" Latin="-5" Hanja="-5" Japanese="-5" Other="-5" Symbol="-5" User="-5"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="4" Height="900" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RATIO Hangul="95" Latin="95" Hanja="95" Japanese="95" Other="95" Symbol="95" User="95"/>
<CHARSPACING Hangul="-5" Latin="-5" Hanja="-5" Japanese="-5" Other="-5" Symbol="-5" User="-5"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="5" Height="1000" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="6" Height="1000" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="2" Latin="2" Hanja="2" Japanese="2" Other="2" Symbol="2" User="2"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="7" Height="1000" TextColor="16711680" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="8" Height="1000" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="9" Height="1000" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHARSPACING Hangul="-50" Latin="-50" Hanja="-50" Japanese="-50" Other="-50" Symbol="-50" User="-50"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="10" Height="1200" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="2" Latin="2" Hanja="2" Japanese="2" Other="2" Symbol="2" User="2"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="11" Height="1200" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="2" Latin="2" Hanja="2" Japanese="2" Other="2" Symbol="2" User="2"/>
<RATIO Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="12" Height="1000" TextColor="16711680" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="13" Height="1100" TextColor="255" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="14" Height="1600" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="2" Latin="2" Hanja="2" Japanese="2" Other="2" Symbol="2" User="2"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="15" Height="1400" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="16" Height="1000" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="true" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="17" Height="1200" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="true" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="18" Height="1400" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="3" Japanese="3" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<BOLD/>
</CHARSHAPE>
<CHARSHAPE Id="19" Height="1200" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="4" Latin="4" Hanja="3" Japanese="3" Other="4" Symbol="4" User="4"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="20" Height="1200" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="3" Japanese="3" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<BOLD/>
</CHARSHAPE>
<CHARSHAPE Id="21" Height="1200" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="4" Latin="4" Hanja="1" Japanese="1" Other="4" Symbol="4" User="4"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="22" Height="1200" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="4" Latin="4" Hanja="1" Japanese="1" Other="4" Symbol="4" User="4"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="23" Height="900" TextColor="16711680" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="24" Height="1200" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="3" Japanese="3" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="25" Height="1000" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="3" Latin="3" Hanja="0" Japanese="0" Other="3" Symbol="3" User="3"/>
<RATIO Hangul="97" Latin="97" Hanja="97" Japanese="97" Other="97" Symbol="97" User="97"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
<CHARSHAPE Id="26" Height="900" TextColor="0" ShadeColor="4294967295" UseFontSpace="false" UseKerning="false" SymMark="0"><FONTID Hangul="2" Latin="2" Hanja="2" Japanese="2" Other="2" Symbol="2" User="2"/>
<RATIO Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHARSPACING Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
<RELSIZE Hangul="100" Latin="100" Hanja="100" Japanese="100" Other="100" Symbol="100" User="100"/>
<CHAROFFSET Hangul="0" Latin="0" Hanja="0" Japanese="0" Other="0" Symbol="0" User="0"/>
</CHARSHAPE>
</CHARSHAPELIST>
<TABDEFLIST Count="3"><TABDEF Id="0" AutoTabLeft="false" AutoTabRight="false"/>
<TABDEF Id="1" AutoTabLeft="true" AutoTabRight="false"/>
<TABDEF Id="2" AutoTabLeft="false" AutoTabRight="true"/>
</TABDEFLIST>
<NUMBERINGLIST Count="1"><NUMBERING Id="1" Start="0"><PARAHEAD Level="1" Alignment="Left" UseInstWidth="true" AutoIndent="true" WidthAdjust="0" TextOffsetType="percent" TextOffset="50" NumFormat="Digit">^1.</PARAHEAD>
<PARAHEAD Level="2" Alignment="Left" UseInstWidth="true" AutoIndent="true" WidthAdjust="0" TextOffsetType="percent" TextOffset="50" NumFormat="HangulSyllable">^2.</PARAHEAD>
<PARAHEAD Level="3" Alignment="Left" UseInstWidth="true" AutoIndent="true" WidthAdjust="0" TextOffsetType="percent" TextOffset="50" NumFormat="Digit">^3)</PARAHEAD>
<PARAHEAD Level="4" Alignment="Left" UseInstWidth="true" AutoIndent="true" WidthAdjust="0" TextOffsetType="percent" TextOffset="50" NumFormat="HangulSyllable">^4)</PARAHEAD>
<PARAHEAD Level="5" Alignment="Left" UseInstWidth="false" AutoIndent="true" WidthAdjust="0" TextOffsetType="percent" TextOffset="50" NumFormat="Digit"></PARAHEAD>
<PARAHEAD Level="6" Alignment="Left" UseInstWidth="false" AutoIndent="true" WidthAdjust="0" TextOffsetType="percent" TextOffset="50" NumFormat="Digit"></PARAHEAD>
<PARAHEAD Level="7" Alignment="Left" UseInstWidth="false" AutoIndent="true" WidthAdjust="0" TextOffsetType="percent" TextOffset="50" NumFormat="Digit"></PARAHEAD>
</NUMBERING>
</NUMBERINGLIST>
<PARASHAPELIST Count="41"><PARASHAPE Id="0" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="1" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-2620" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="1" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="true"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="2" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="3" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="3000" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="4" Align="Justify" VerAlign="Baseline" HeadingType="Outline" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="20" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="2000" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="5" Align="Justify" VerAlign="Baseline" HeadingType="Outline" Heading="0" Level="1" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="20" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="4000" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="6" Align="Justify" VerAlign="Baseline" HeadingType="Outline" Heading="0" Level="2" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="20" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="6000" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="7" Align="Justify" VerAlign="Baseline" HeadingType="Outline" Heading="0" Level="3" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="20" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="8000" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="8" Align="Justify" VerAlign="Baseline" HeadingType="Outline" Heading="0" Level="4" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="20" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="10000" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="9" Align="Justify" VerAlign="Baseline" HeadingType="Outline" Heading="0" Level="5" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="20" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="12000" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="10" Align="Justify" VerAlign="Baseline" HeadingType="Outline" Heading="0" Level="6" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="20" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="14000" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="11" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="2" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="150"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="12" Align="Right" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="13" Align="Right" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="100"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="14" Align="Right" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="15" Align="Center" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="16" Align="Center" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="17" Align="Right" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="180"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="18" Align="Center" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="2" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="150"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="19" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="2" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="150"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="20" Align="Right" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="2" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="150"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="21" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-4600" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="22" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-2400" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="23" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-1936" Left="0" Right="0" Prev="1000" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="24" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-2336" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="25" Align="Center" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="26" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-3904" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="27" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-2400" Left="0" Right="0" Prev="1400" Next="0" LineSpacingType="Percent" LineSpacing="180"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="28" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-4600" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="180"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="29" Align="Right" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="2296" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="180"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="30" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-7800" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="31" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-10400" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="32" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="2000" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="33" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="4000" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="160"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="34" Align="Center" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="0" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="35" Align="Center" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="false" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-1000" Left="11496" Right="12552" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="36" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-1900" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="37" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-3600" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="38" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-6500" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="39" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-8000" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
<PARASHAPE Id="40" Align="Justify" VerAlign="Baseline" HeadingType="None" Heading="0" Level="0" TabDef="0" BreakLatinWord="KeepWord" BreakNonLatinWord="true" Condense="0" WidowOrphan="false" KeepWithNext="false" KeepLines="false" PageBreakBefore="false" FontLineHeight="false" SnapToGrid="false"><PARAMARGIN Indent="-10500" Left="0" Right="0" Prev="0" Next="0" LineSpacingType="Percent" LineSpacing="130"/>
<PARABORDER BorderFill="1" OffsetLeft="0" OffsetRight="0" OffsetTop="0" OffsetBottom="0" Connect="false" IgnoreMargin="false"/>
</PARASHAPE>
</PARASHAPELIST>
<STYLELIST Count="14"><STYLE Id="0" Type="Para" Name="바탕글" EngName="Normal" ParaShape="2" CharShape="1" NextStyle="0" LangId="1042"/>
<STYLE Id="1" Type="Para" Name="본문" EngName="Body" ParaShape="3" CharShape="1" NextStyle="1" LangId="1042"/>
<STYLE Id="2" Type="Para" Name="개요 1" EngName="Outline 1" ParaShape="4" CharShape="1" NextStyle="2" LangId="1042"/>
<STYLE Id="3" Type="Para" Name="개요 2" EngName="Outline 2" ParaShape="5" CharShape="1" NextStyle="3" LangId="1042"/>
<STYLE Id="4" Type="Para" Name="개요 3" EngName="Outline 3" ParaShape="6" CharShape="1" NextStyle="4" LangId="1042"/>
<STYLE Id="5" Type="Para" Name="개요 4" EngName="Outline 4" ParaShape="7" CharShape="1" NextStyle="5" LangId="1042"/>
<STYLE Id="6" Type="Para" Name="개요 5" EngName="Outline 5" ParaShape="8" CharShape="1" NextStyle="6" LangId="1042"/>
<STYLE Id="7" Type="Para" Name="개요 6" EngName="Outline 6" ParaShape="9" CharShape="1" NextStyle="7" LangId="1042"/>
<STYLE Id="8" Type="Para" Name="개요 7" EngName="Outline 7" ParaShape="10" CharShape="1" NextStyle="8" LangId="1042"/>
<STYLE Id="9" Type="Para" Name="쪽 번호" EngName="Page Number" ParaShape="2" CharShape="0" NextStyle="9" LangId="1042"/>
<STYLE Id="10" Type="Para" Name="머리말" EngName="Header" ParaShape="11" CharShape="2" NextStyle="10" LangId="1042"/>
<STYLE Id="11" Type="Para" Name="각주" EngName="Footnote" ParaShape="0" CharShape="3" NextStyle="11" LangId="1042"/>
<STYLE Id="12" Type="Para" Name="미주" EngName="Endnote" ParaShape="0" CharShape="3" NextStyle="12" LangId="1042"/>
<STYLE Id="13" Type="Para" Name="메모" EngName="Memo" ParaShape="1" CharShape="4" NextStyle="13" LangId="1042"/>
</STYLELIST>
</MAPPINGTABLE>
</HEAD>
<BODY><SECTION Id="0"><P ParaShape="34"><TEXT CharShape="10"><SECDEF SpaceColumns="1134" TabStop="8000" OutlineShape="1"><STARTNUMBER PageStartsOn="Both" Page="0" Figure="0" Table="0" Equation="0"/>
<HIDE Header="false" Footer="false" MasterPage="false" Border="false" Fill="false" PageNumPos="false" EmptyLine="false"/>
<PAGEDEF Landscape="0" Width="59528" Height="84188" GutterType="LeftOnly"><PAGEMARGIN Left="4252" Right="4252" Top="5669" Bottom="4252" Header="3600" Footer="3600" Gutter="0"/>
</PAGEDEF>
<FOOTNOTESHAPE><AUTONUMFORMAT Type="Digit" SuffixChar=")" Superscript="false"/>
<NOTELINE Length="5cm" Type="Solid" Width="0.12mm" Color="0"/>
<NOTESPACING AboveLine="850" BelowLine="567" BetweenNotes="283"/>
<NOTENUMBERING Type="Continuous"/>
<NOTEPLACEMENT Place="EachColumn" BeneathText="false"/>
</FOOTNOTESHAPE>
<ENDNOTESHAPE><AUTONUMFORMAT Type="Digit" SuffixChar=")" Superscript="false"/>
<NOTELINE Length="14692344" Type="Solid" Width="0.12mm" Color="0"/>
<NOTESPACING AboveLine="850" BelowLine="567" BetweenNotes="0"/>
<NOTENUMBERING Type="Continuous"/>
<NOTEPLACEMENT Place="EndOfDocument" BeneathText="false"/>
</ENDNOTESHAPE>
<PAGEBORDERFILL Type="Both" BorferFill="0" TextBorder="true" HeaderInside="false" FooterInside="false" FillArea="Paper"><PAGEOFFSET Left="1417" Right="1417" Top="1417" Bottom="1417"/>
</PAGEBORDERFILL>
<PAGEBORDERFILL Type="Even" BorferFill="0" TextBorder="true" HeaderInside="false" FooterInside="false" FillArea="Paper"><PAGEOFFSET Left="1417" Right="1417" Top="1417" Bottom="1417"/>
</PAGEBORDERFILL>
<PAGEBORDERFILL Type="Odd" BorferFill="0" TextBorder="true" HeaderInside="false" FooterInside="false" FillArea="Paper"><PAGEOFFSET Left="1417" Right="1417" Top="1417" Bottom="1417"/>
</PAGEBORDERFILL>
</SECDEF>
<COLDEF Type="Newspaper" Count="1" Layout="Left" SameSize="true"/>
<PICTURE><SHAPEOBJECT InstId="1517942637" ZOrder="14" NumberingType="Figure" TextWrap="BehindText" Lock="false"><SIZE Width="5556" Height="5556" WidthRelTo="Absolute" HeightRelTo="Absolute" Protect="false"/><POSITION TreatAsChar="false" VertRelTo="Para" HorzRelTo="Column" VertAlign="Top" HorzAlign="Left" VertOffset="-901" HorzOffset="45187" FlowWithText="false" AllowOverlap="true"/><OUTSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/></SHAPEOBJECT><SHAPECOMPONENT XPos="-2647" YPos="-2016" GroupLevel="0" OriWidth="10208" OriHeight="12300" CurWidth="10681" CurHeight="12870" HorzFlip="false" VertFlip="false"><ROTATIONINFO Angle="0" CenterX="2778" CenterY="2778"/><RENDERINGINFO><TRANSMATRIX E1="1.00" E2="0.00" E3="-2647.00" E4="0.00" E5="1.00" E6="-2016.00"/><SCAMATRIX E1="1.05" E2="0.00" E3="0.00" E4="0.00" E5="1.05" E6="0.00"/><ROTMATRIX E1="1.04" E2="0.00" E3="-102.42" E4="0.00" E5="1.04" E6="-76.05"/><SCAMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><SCAMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><SCAMATRIX E1="1.01" E2="0.00" E3="-22.70" E4="0.00" E5="1.01" E6="-15.04"/><ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><SCAMATRIX E1="0.75" E2="0.00" E3="0.00" E4="0.00" E5="0.75" E6="0.00"/><ROTMATRIX E1="1.19" E2="0.00" E3="-494.20" E4="0.00" E5="1.19" E6="-374.90"/><SCAMATRIX E1="0.80" E2="0.00" E3="-166.00" E4="0.00" E5="0.80" E6="2941.00"/><ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="-0.45"/><SCAMATRIX E1="0.82" E2="0.00" E3="-17.00" E4="0.00" E5="0.79" E6="0.00"/><ROTMATRIX E1="1.21" E2="0.00" E3="-927.59" E4="0.00" E5="1.45" E6="227.53"/><SCAMATRIX E1="0.70" E2="0.00" E3="5430.00" E4="0.00" E5="0.73" E6="-584.00"/><ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="0.69" E6="-195.65"/></RENDERINGINFO></SHAPECOMPONENT><LINESHAPE Color="11866128" Width="85" Style="Solid" EndCap="Round" OutlineStyle="Outer"/><IMAGERECT X0="0" Y0="0" X1="10208" Y1="0" X2="10208" Y2="12300" X3="0" Y3="12300"/><IMAGECLIP Left="567" Top="567" Right="14433" Bottom="14433"/><INSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/><IMAGE Bright="0" Contrast="0" Effect="RealPic" BinItem="9"/></PICTURE>
<CHAR></CHAR>
</TEXT>
</P>
<P ParaShape="35"><TEXT CharShape="10"><CHAR>의료기기 허가·신고·심사 등에 관한 규정</CHAR>
</TEXT>
</P>
<P ParaShape="16"><TEXT CharShape="12"><HEADER ApplyPageType="Both" TextWidth="48190" TextHeight="2835" HasTextRef="false" HasNumRef="false"><PARALIST TextDirection="0" LineWrap="Break" VertAlign="Top"><P ParaShape="13"><TEXT CharShape="8"><TABLE PageBreak="Cell" RepeatHeader="true" RowCount="1" ColCount="1" CellSpacing="0" BorderFill="2"><SHAPEOBJECT InstId="1201155746" ZOrder="0" NumberingType="Table" Lock="false"><SIZE Width="50929" Height="1982" WidthRelTo="Absolute" HeightRelTo="Absolute" Protect="false"/>
<POSITION TreatAsChar="true" AffectLSpacing="false" VertAlign="Top" HorzAlign="Left" VertOffset="0" HorzOffset="0" FlowWithText="true" HoldAnchorAndSO="false"/>
<OUTSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/>
</SHAPEOBJECT>
<INSIDEMARGIN Left="510" Right="510" Top="141" Bottom="141"/>
<ROW><CELL ColAddr="0" RowAddr="0" ColSpan="1" RowSpan="1" Width="50929" Height="1982" Header="false" HasMargin="false" Protect="false" Editable="false" Dirty="false" BorderFill="4"><PARALIST TextDirection="0" LineWrap="Break" VertAlign="Center"><P ParaShape="13"><TEXT CharShape="5"><CHAR>의료기기 허가·신고·심사 등에 관한 규정</CHAR>
</TEXT>
</P>
</PARALIST>
</CELL>
</ROW>
</TABLE>
<CHAR></CHAR>
</TEXT>
</P>
<P ParaShape="12"><TEXT CharShape="9"/>
</P>
</PARALIST>
</HEADER>
<FOOTER ApplyPageType="Both" TextWidth="48190" TextHeight="2835" HasTextRef="false" HasNumRef="false"><PARALIST TextDirection="0" LineWrap="Break" VertAlign="Bottom"><P ParaShape="14"><TEXT CharShape="9"/>
</P>
<P ParaShape="15"><TEXT CharShape="9"><TABLE PageBreak="Cell" RepeatHeader="true" RowCount="1" ColCount="3" CellSpacing="0" BorderFill="2"><SHAPEOBJECT InstId="1201155747" ZOrder="1" NumberingType="Table" Lock="false"><SIZE Width="51024" Height="1982" WidthRelTo="Absolute" HeightRelTo="Absolute" Protect="false"/>
<POSITION TreatAsChar="true" AffectLSpacing="false" VertAlign="Top" HorzAlign="Left" VertOffset="0" HorzOffset="0" FlowWithText="true" HoldAnchorAndSO="false"/>
<OUTSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/>
</SHAPEOBJECT>
<INSIDEMARGIN Left="510" Right="510" Top="141" Bottom="141"/>
<ROW><CELL ColAddr="0" RowAddr="0" ColSpan="1" RowSpan="1" Width="12623" Height="1982" Header="false" HasMargin="false" Protect="false" Editable="false" Dirty="false" BorderFill="3"><PARALIST TextDirection="0" LineWrap="Break" VertAlign="Center"><P ParaShape="19" Style="10"><TEXT CharShape="5"><PICTURE><SHAPEOBJECT InstId="1201155748" ZOrder="2" NumberingType="Figure" Lock="false"><SIZE Width="1828" Height="1131" WidthRelTo="Absolute" HeightRelTo="Absolute" Protect="false"/>
<POSITION TreatAsChar="true" AffectLSpacing="false" VertAlign="Top" HorzAlign="Left" VertOffset="0" HorzOffset="0" FlowWithText="true"/>
<OUTSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/>
</SHAPEOBJECT>
<SHAPECOMPONENT XPos="0" YPos="-2469" GroupLevel="0" OriWidth="23760" OriHeight="14700" CurWidth="22847" CurHeight="14137" HorzFlip="false" VertFlip="false"><ROTATIONINFO Angle="0" CenterX="914" CenterY="565"/>
<RENDERINGINFO><TRANSMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="-2469.00"/>
<SCAMATRIX E1="0.96" E2="0.00" E3="0.00" E4="0.00" E5="0.96" E6="94.48"/>
<ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/>
<SCAMATRIX E1="0.08" E2="0.00" E3="0.00" E4="0.00" E5="0.08" E6="2469.00"/>
<ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/>
</RENDERINGINFO>
</SHAPECOMPONENT>
<IMAGERECT X0="0" Y0="0" X1="23760" Y1="0" X2="23760" Y2="14700" X3="0" Y3="14700"/>
<IMAGECLIP Left="0" Top="0" Right="23700" Bottom="14640"/>
<INSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/>
<IMAGE Bright="0" Contrast="0" Effect="RealPic" BinItem="3"/>
</PICTURE>
<CHAR>&nbsp;법제처</CHAR>
</TEXT>
<TEXT CharShape="5"><CHAR></CHAR>
</TEXT>
</P>
</PARALIST>
</CELL>
<CELL ColAddr="1" RowAddr="0" ColSpan="1" RowSpan="1" Width="25701" Height="1982" Header="false" HasMargin="false" Protect="false" Editable="false" Dirty="false" BorderFill="3"><PARALIST TextDirection="0" LineWrap="Break" VertAlign="Center"><P ParaShape="18" Style="10"><TEXT CharShape="26"><CHAR>-&nbsp;</CHAR>
<AUTONUM Number="1" NumberType="Page"><AUTONUMFORMAT Type="Digit" Superscript="false"/>
</AUTONUM>
<CHAR>&nbsp;/&nbsp;</CHAR>
<AUTONUM Number="1" NumberType="TotalPage"><AUTONUMFORMAT Type="Digit" Superscript="false"/>
</AUTONUM>
<CHAR>&nbsp;-</CHAR>
</TEXT>
</P>
</PARALIST>
</CELL>
<CELL ColAddr="2" RowAddr="0" ColSpan="1" RowSpan="1" Width="12623" Height="1982" Header="false" HasMargin="false" Protect="false" Editable="false" Dirty="false" BorderFill="3"><PARALIST TextDirection="0" LineWrap="Break" VertAlign="Center"><P ParaShape="20" Style="10"><TEXT CharShape="5"><PICTURE><SHAPEOBJECT InstId="1310949373" ZOrder="3" NumberingType="Figure" Lock="false"><SIZE Width="1417" Height="1417" WidthRelTo="Absolute" HeightRelTo="Absolute" Protect="false"/>
<POSITION TreatAsChar="true" AffectLSpacing="false" VertAlign="Top" HorzAlign="Left" VertOffset="0" HorzOffset="0" FlowWithText="true"/>
<OUTSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/>
</SHAPEOBJECT>
<SHAPECOMPONENT XPos="0" YPos="-887" GroupLevel="0" OriWidth="2760" OriHeight="2760" CurWidth="1416" CurHeight="1416" HorzFlip="false" VertFlip="false"><ROTATIONINFO Angle="0" CenterX="708" CenterY="708"/>
<RENDERINGINFO><TRANSMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="-887.00"/>
<SCAMATRIX E1="0.51" E2="0.00" E3="0.00" E4="0.00" E5="0.51" E6="887.00"/>
<ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/>
</RENDERINGINFO>
</SHAPECOMPONENT>
<IMAGERECT X0="0" Y0="0" X1="2760" Y1="0" X2="2760" Y2="2760" X3="0" Y3="2760"/>
<IMAGECLIP Left="0" Top="0" Right="2700" Bottom="2700"/>
<INSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/>
<IMAGE Bright="0" Contrast="0" Effect="RealPic" BinItem="2"/>
</PICTURE>
<CHAR>&nbsp;국가법령정보센터</CHAR>
</TEXT>
</P>
</PARALIST>
</CELL>
</ROW>
</TABLE>
<CHAR></CHAR>
</TEXT>
</P>
</PARALIST>
</FOOTER>
</TEXT>
<TEXT CharShape="12"><CHAR>[시행 2017.7.10] [식품의약품안전처고시 제2017-58호, 2017.7.10, 일부개정]</CHAR>
</TEXT>
</P>
<P ParaShape="16"><TEXT CharShape="6"/></P>
<P ParaShape="17"><TEXT CharShape="25"><CHAR>식품의약품안전처(의료기기정책과), 043-719-3760</CHAR></TEXT></P>
<P ParaShape="2"><TEXT CharShape="5"/>
</P>
{전문}
<P ParaShape="2"><TEXT CharShape="6"><CHAR>        제1장 총칙</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제1조(목적)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 이 규정은 「의료기기법」제8조 및 제6조의3,「의료기기법 시행규칙」 제5조, 제6조, 제7조, 제9조, 제25조, 제26조, 제27조, 제30조, 제33조, 제38조, 제41조, 제46조, 제47조, 「첨단의료복합단지 지정 및 지원에 관한 특별법」 제24조제3항 및 「첨단의료복합단지 지정 및 지원에 관한 특별법 시행규칙」 제5조제6항에 따른 제조·수입 의료기기의 허가·인증·신고 등에 관한 세부사항 및 의료기기 기술문서 등의 심사에 필요한 세부사항, 검사필증 발행 절차 등에 관한 사항, 사전검토의 방법 및 절차 등의 세부사항, 판매업 신고 등이 면제되는 의료기기, 전시할 목적의 의료기기를 승인받고자 하는 자의 신청방법과 절차, 의료기기의 성능 개선 허용 범위, 희소의료기기를 지정하기 위한 지정기준과 절차 및 첨단의료복합단지 입주의료연구개발기관의 제조 허가·인증 신청과 수입 승인 신청에 대한 수수료 감면 등을 정함으로써 의료기기 허가 등 관리의 적정을 기함을 목적으로 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제2조(정의)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 이 규정에서 사용하는 용어의 뜻은 다음과 같다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. "동일제품군"이란 제조국, 제조사, 품목명이 동일한 의료기기 중 사용목적, 사용방법, 제조방법 및 색소나 착향제를 제외한 원재료(기구·기계는 제외한다)가 동일한 것으로 색상, 치수 등의 차이가 있거나 구성 부분품이 변경 또는 추가되는 일련의 모델(시리즈 제품)들로 구성된 제품군을 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. "일회용 의료기기"란 한 번 사용할 목적 또는 한 번의 시술과정에서 한 환자에게 사용할 목적인 의료기기를 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. "조합의료기기"란 2가지 이상의 의료기기가 모여 하나의 의료기기가 되는 것으로서 복합적인 기능을 발휘하는 의료기기를 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. "한벌구성의료기기"란 2가지 이상의 의료기기를 하나의 포장단위로 구성한 의료기기를 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. "첨단의료기기"란 융·복합의료기기를 포함하여 바이오기술, 정보기술, 나노기술, 로봇기술, 의료기술 등 혁신적 과학기술을 기반으로 한 새로운 의료기기를 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  6. "부분품"이란 의료기기의 본체를 구성하기 위하여 필요한 부분을 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  7. "시험규격"이란 해당 제품의 안전성 및 성능을 검증하기 위하여 설정하는 시험항목, 시험기준 및 시험방법을 말한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  8. "품목"이란 「의료기기법 시행규칙」(이하 "시행규칙"이라 한다) 별표 1 및 「의료기기 품목 및 품목별 등급에 관한 규정」(식품의약품안전처 고시)의 소분류에 해당하는 개별 제품을 말한다. 다만, 소분류에 해당되지 않아 중분류한 품목은 중분류에 해당하는 개별 제품을 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  9. "품목류"란 시행규칙 별표 1 및 「의료기기 품목 및 품목별 등급에 관한 규정」에 따른 소분류를 말한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  10. "새로운제품"이란 이미 허가·인증 받은 의료기기와 사용목적, 작용원리 또는 원재료(의료용품에 한한다) 등이 동등하지 아니한 의료기기를 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  11. "개량제품"이란 이미 허가·인증 받은 의료기기와 사용목적, 작용원리, 원재료(의료용품에 한한다)는 동등하나 성능, 시험규격, 사용방법 등이 동등하지 아니한 의료기기를 말한다. 다만, 체외진단용 의료기기는 이미 허가·인증 받은 의료기기와 사용목적, 작용원리는 동등하나 원재료 또는 성능이 동등하지 아니한 제품을 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  12. "동등제품"이란 이미 허가·인증 받은 의료기기와 사용목적, 작용원리, 원재료(의료용품에 한한다), 성능, 시험규격 및 사용방법 등이 동등한 의료기기를 말한다. 다만, 체외진단용 의료기기는 사용목적, 작용원리, 원재료 및 성능이 동등한 경우를 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  13. "체외진단용 의료기기"란 인체에서 유래한 시료를 검체로 하여 검체 중의 물질을 검사하여 질병 진단, 예후 관찰, 혈액 또는 조직 적합성 판단 등의 정보 제공을 목적으로 체외에서 사용되는 시약을 말한다. 다만, 실험실에서 조제하여 사용하는 조제시약은 제외한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  14. "사용기간 또는 유효기간"이란 제조자가 의도한 의료기기의 사용목적 대로 작용할 수 있도록 성능 및 안전성(멸균의료기기의 경우 멸균)등이 유지되는 실제시간을 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  15. "국제표준화기술문서"란  의료기기 규제당국자포럼(IMDRF, International Medical Device Regulators Forum)에서 국가간 의료기기 규제차이를 없애기 위해 개발한 표준화된 문서로 기술문서 개요(STED, Summary Technical Documentation)와 이에 대한 첨부자료로 구성된다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  16. "인체이식형 의료기기"란 인체에 30일 이상 연속적으로 유지되는 것을 목적으로 삽입하는 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  17. "단계별 심사"란 법 제11조 및 시행규칙 제9조에 따라 제조허가 신청 전에 허가에 필요한 기술문서 등의 심사 자료에 대하여 미리 제품 개발 단계별로 나누어 심사하는 것을 말한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제2장 의료기기 품목 및 품목류 허가·인증·신고</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제3조(의료기기 허가·인증·신고의 신청 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 품목류 제조·수입 인증을 받거나 신고를 하여야 하는 의료기기는 별표 1과 같으며, 이 경우 동일제품군에 해당하는 대표 제품 하나 이상을 대상으로 신청하여야 한다. 다만, 신청하고자 하는 제품 중 구조·원리·성능·사용목적 및 사용방법 등이 본질적으로 동등하지 아니하여 시행규칙 제9조제2항제6호에 따른 임상시험에 관한 자료를 제출하여야 하는 경우에는 별도의 품목허가로 신청하여야 한다. </CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 품목별로 허가·인증을 받거나 신고하려는 자는 신청하고자 하는 제품이 동일제품군에 해당하는 경우에 하나의 품목허가·인증 또는 품목신고로 신청하여야 한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 조합의료기기의 품목명은 주된 기능을 발휘하는 의료기기로 기재하며, 조합된 각 의료기기의 등급이 다를 경우에는 그 의료기기 중 가장 높은 위해도의 등급으로 기재한다. 다만, 조합된 각 의료기기를 분리할 경우에는 각각의 의료기기에 대해 품목허가·인증을 받거나 신고를 하여야 한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ④ 한벌구성의료기기는 다음 각 호에 따라 품목허가·인증을 받거나 신고를 하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 품목명은 한벌 구성된 의료기기의 주된 사용목적 또는 기능을 발휘하는 의료기기로 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 한벌 구성된 각각의 의료기기 등급이 다를 경우에는 그 의료기기 중 가장 높은 위해도의 등급으로 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 품목허가·인증을 받거나 신고 시에는 한벌 구성된 각각의 의료기기에 대한 분류번호, 명칭(제품명, 품목명, 모델명), 등급을 모두 기재하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑤ 제4항에 따라 이미 허가·인증을 받은 한벌구성의료기기를 각각의 의료기기로 품목허가·인증을 받고자 하는 경우에는 기술문서와 임상시험자료 등(이하 "기술문서 등"이라 한다) 또는 기술문서 등의 심사결과통지서를 제출하지 아니할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑥ 수출만을 목적으로 하는 의료기기를 제조하려는 자라도 식품의약품안전처장(이하 "식약처장"이라 한다)에게 제조허가를 신청하거나 법 제42조에 따라 지정받은 의료기기정보기술지원센터의 장(이하 "의료기기정보기술지원센터장"이라 한다)에게 제조인증·신고를 신청할 수 있다. 이 경우 허가·인증·신고를 신청하려는 자는 시행규칙 제5조제1항에 따른 서류를 제출하지 아니하고 허가·인증·신고를 신청할 수 있다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑦ 제6항에 따라 허가·인증을 받은 품목을 국내에 판매하고자 할 경우에는 별도의 제조·수입허가·인증을 받아야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑧ 신개발의료기기 등과 같이 분류결정 등에 장시간 소요되는 제품에 대하여는 중분류명 또는 별도로 정한 품목명과 분류번호를 사용하여 품목허가·인증을 하거나 신고를 수리할 수 있다. 이 경우 시행규칙 제2조에 따른 등급분류기준을 적용하여 등급을 분류한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑨ 식약처장은 2등급 의료기기 중 동등제품으로 3회 이상 허가·인증 받은 제품에 대하여 사용목적, 작용원리, 원재료, 성능, 시험규격 및 사용방법 등을 식품의약품안전처 홈페이지를 통해 공고할 수 있으며, 이를 추가 또는 변경하거나 제외할 경우에는 변경 공고하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑩ 제9항에 따라 공고된 제품(이하 "동등공고제품"이라 한다)의 경우, 시행규칙 제5조제1항제2호에 따른 서류를 갈음하여 동등공고제품에 적합함을 증명하는 안전성 및 성능에 관한 시험성적서를 제출하여야 한다. 이 경우 시험성적서는 「식품·의약품분야 시험·검사 등에 관한 법률」제6조제2항제4호에 따라 식약처장이 지정한 의료기기 시험·검사기관(이하 "시험·검사기관"이라 한다)이 발행한 것에 한한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑪ 이미 허가·인증받은 품목과 동일제품임을 확인받고자 하는 자는 다음 각 호의 요건에 해당함을 입증할 수 있는 자료를 첨부하여 식약처장에게 신청하여야 하며, 식약처장은 검토사항에 대한 결과를 10일 이내에 통지하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 이미 허가·인증받은 의료기기와 사용목적, 작용원리, 원재료, 성능, 시험규격 및 사용방법 등이 동일한 의료기기 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 동일 제조소(수입의료기기의 경우 제조국가·제조회사·제조소가 동일한 경우를 말한다)에서 제조된 의료기기 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 제조의뢰자로부터 위탁받은 제조자가 제품을 설계·개발·생산하는 방식으로 제조한 의료기기(국내에서 제조된 의료기기에 한한다) </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑫ 첨단의료복합단지 입주의료연구개발기관이 의료기기의 제조허가·인증을 신청하는 경우에는 「첨단의료복합단지 지정 및 지원에 관한 특별법 시행규칙」 제5조제6항에 따라 준용되는 같은법 시행규칙 제4조제6항에 따라 시행규칙 별표 10에 따른 수수료 전부를 감면한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑬ 첨단의료복합단지 입주의료연구개발기관이 「첨단의료복합단지 지원에 관한 특별법」 제24조제2항에 따라 수입승인을 신청하는 경우에는 수수료를 면제한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑭ 첨단의료기기 등의 제조허가를 받고자 하는 자는 시행규칙 제9조에 따른 기술문서 등의 심사를 개발 단계별로 신청할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제4조(제조·수입 신고의 처리 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 의료기기 제조 또는 수입신고를 하려는 자가 의료기기정보기술지원센터장에게 제출한 의료기기 제조(수입) 신고서(이하 "신고서"라 한다)가 의료기기정보기술지원센터 전자민원시스템에 등록이 완료된 경우에는 신고가 수리된 것으로 본다. 다만, 다음 각 호의 어느 하나에 해당하는 경우에는 그러하지 아니하다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 시행규칙 제4조에 따른 허가·인증 대상 의료기기를 신고서로 제출한 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 의료기기에 해당하지 않는 제품을 신고한 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 제5조 각 호의 어느 하나에 해당하는 원자재를 사용하거나 함유한 의료기기를 신고한 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 제1항에도 불구하고 의료기기 제조(수입)업 허가신청서와 동시에 제출된 신고서의 경우에는 제조·수입업 허가 시 수리된 것으로 본다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제5조(안전성·유효성 문제 원자재 사용 의료기기 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 법 제6조의3제1항제2호에 따른 안전성·유효성에 문제가 있는 원자재 또는 의료기기는 다음 각 호와 같다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 수은(Hg)(치과용 제외)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 석면</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 디에틸헥실프탈레이트(di-(2- ethylhexyl) -phthalate, DEHP), 디부틸프탈레이트(Dibutylphthalate, DBP) 또는 벤질부틸프탈레이트(Benzyl butyl phthalate, BBP) 등 프탈레이트류 함유 수액세트</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제6조(제조·수입허가·인증신청서 및 신고서의 작성 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 의료기기 제조(수입)허가·인증 신청서(이하 "허가·인증신청서"라 한다) 또는 신고서는 이 규정에서 정하는 첨부서류 등을 근거로 적합하게 작성하여야 하고, 허가·인증신청서 및 신고서 항목은 제8조부터 제18조까지의 규정에 따라 기재하여야 한다. 이 경우 식약처장이 정한 전용프로그램을 사용하여 작성하거나, 전자적 기록매체(CD, 디스켓 등)에 수록하여 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 수출만을 목적으로 하는 의료기기의 경우에는 허가·인증신청서 또는 신고서 비고란에 "수출용에 한함"이라는 표기를 하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 품목류 인증·신고 대상 의료기기의 경우에는 인증신청서 또는 신고서의 "품목류" 해당란에 해당 표기를 하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ④ 중고의료기기의 수입허가·인증·신고 신청 시 허가·인증신청서 또는 신고서 비고란에 "중고의료기기"라는 표기를 하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑤ 동등제품의 경우 허가·인증신청서 비고란에 "동등제품(동등비교 제품 허가번호)"이라는 표기를 하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑥ 동등공고제품 의료기기의 경우 허가·인증 신청서 비고란에 "동등공고제품(공고번호, 품목명, 분류번호)"이라는 표기를 하여야 한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑦ 인체이식형 의료기기의 경우 허가·인증신청서 비고란에 "인체이식형 의료기기"라는 표기를 하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑧ 「첨단의료복합단지 지정 및 지원에 관한 특별법」제24조제3항에 따라 입주의료연구개발기관이 제1항의 허가·인증신청서를 작성하는 경우에는 영어로 작성하여 제출할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제7조(허가·인증신청서 및 신고서 항목)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 허가·인증신청서에 기재하여야 하는 항목은 다음 각 호와 같다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 명칭(제품명, 품목명, 모델명)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 분류번호(등급)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 모양 및 구조</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 원재료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. 제조방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  6. 성능</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  7. 사용목적</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  8. 사용방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  9. 사용 시 주의사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  10. 포장단위</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  11. 저장방법 및 사용기간</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  12. 시험규격</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  13. 제조원(수입 또는 제조공정 전부 위탁의 경우)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  14. 허가(인증)조건</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  15. 비고</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 신고서에 기재하여야 하는 항목은 다음 각 호와 같다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 명칭(제품명, 품목명, 모델명)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 분류번호(등급)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 모양 및 구조</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 성능</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. 사용목적</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  6. 사용방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  7. 사용 시 주의사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  8. 제조원(수입 또는 제조공정 전부 위탁의 경우)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  9. 비고</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제3장 의료기기 허가·인증신청서 및 신고서 항목 기재 세부사항 등</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제8조(명칭)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 의료기기의 명칭은 다음 각 호의 어느 하나에 따라 기재하여야 한다. 다만, 품목류 인증·신고 시에는 신청한 대표 제품의 모델명에 덧붙여 "등 동일제품군"이라는 문구를 기재한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 제품명을 기재하는 경우에는 "제조(수입)업소명·제품명", "품목명", "모델명"을 각각 기재한다. 이때, 제조(수입)업소명은 생략할 수 있고 제품명은 두 개 이상 인정한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제품명을 기재하지 아니하는 경우에는 "제조(수입)업소명·품목명", "모델명"을 각각 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 제품명은 이미 허가·인증을 받거나 신고한 의료기기의 제품명과 동일하여서는 아니 된다. 다만, 다음 각 호의 어느 하나에 해당하는 경우에는 그러하지 아니하다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 허가·인증·신고가 취소된 의료기기와 사용목적, 작용원리 및 원재료 등이 동일한 의료기기로서 취소된 날부터 1년이 지난 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 동일한 제조(수입)업자가 허가·인증·신고 취하 후 동일한 제품을 허가·인증·신고하는 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 서로 다른 수입업자가 제조원이 같은 동일한 제품을 수입하는 경우에 수입업소명을 병기하여 구분하는 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 제1항 각 호의 품목명은 「의료기기 품목 및 품목별 등급에 관한 규정」에 따라 분류된 품목 중 어느 하나에 해당할 경우 그 품목명에 품목분류번호 및 등급을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ④ 조합의료기기 및 한벌구성의료기기의 경우에는 주된 사용목적 및 상위등급에 따라 각각의 의료기기별로 제1항부터 제3항까지의 규정에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑤ 제2항의 본문에도 불구하고 이미 허가·인증을 받거나 신고한 그 제품과 유사한 사용목적에 해당하는 품목의 경우에는 허가·인증받은 제품의 제품명에 문자, 단어 또는 숫자 등을 덧붙이거나 교체한 제품명(예: △△-에이 디 에스 등, △△-2)을 기재할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑥ 수출명을 따로 기재할 필요가 있는 경우에는 "수출명: ○○○○"의 형식으로 괄호 안에 병기한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제9조(모양 및 구조)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 모양 및 구조는 다음 각 호에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 해당 제품의 작용원리를 포함하여 모양·구조·중량 및 치수 등을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제1호에도 불구하고 전기·기계적 원리를 이용하는 의료기기의 경우에는 다음 각 목에 해당하는 내용을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 모양·구조·중량·치수 및 각 부분의 기능</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 전기적 정격</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 전기충격에 대한 보호형식 및 보호정도</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 안전장치</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    마. 작동계통도 및 작동계통도에 따른 작동원리</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    바. 전기·기계적 안전성을 검증 할 수 있는 절연부의 전기회로도(전원부, 장착부 등을 포함한다) 또는 전기절연도(Insulation Diagram)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    사. 소프트웨어의 구조 및 주요기능</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 제1호에도 불구하고 소프트웨어가 단독으로 사용되는 경우에는 다음 각 목에 해당하는 내용을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 모양·구조 및 각 부분의 기능</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 소프트웨어의 구조 및 주요기능</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 한벌구성의료기기의 경우에는 각각의 의료기기의 목록을 작성하고 제1호·제2호 또는 제3호의 규정에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제10조(원재료)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 원재료는 다음 각 호에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 전기를 사용하지 않는 기구·기계, 의료용품, 치과재료에 해당 하는 경우에는 다음 각 목에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 원재료는 다음 표에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>&nbsp;&nbsp;&nbsp;</CHAR><PICTURE><SHAPEOBJECT InstId="1517942632" ZOrder="8" NumberingType="Figure" TextFlow="BothSides" Lock="false"><SIZE Width="44916" Height="6916" WidthRelTo="Absolute" HeightRelTo="Absolute" Protect="false"/><POSITION TreatAsChar="true" AffectLSpacing="false" VertAlign="Top" HorzAlign="Left" VertOffset="0" HorzOffset="0" FlowWithText="true"/><OUTSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/></SHAPEOBJECT><SHAPECOMPONENT XPos="0" YPos="0" GroupLevel="0" OriWidth="5910" OriHeight="910" HorzFlip="false" VertFlip="false"><ROTATIONINFO Angle="0" CenterX="4960" CenterY="480"/><RENDERINGINFO><TRANSMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><SCAMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/></RENDERINGINFO></SHAPECOMPONENT><IMAGERECT X0="0" Y0="0" X1="5910" Y1="0" X2="5910" Y2="910" X3="0" Y3="910"/><IMAGECLIP Left="0" Top="0" Right="0" Bottom="0"/><INSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/><IMAGE Bright="0" Contrast="0" Effect="RealPic" BinItem="4"/></PICTURE><CHAR> </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 부분품의 명칭란에는 해당 의료기기를 구성하는 부분품별 명칭을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 원재료명 또는 성분명란에는 해당 의료기기의 부분품을 구성하는 각 원재료의 일반명 또는 화학명을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 규격란에는 원재료에 대한 규격이 있는 경우에는 해당 규격(KS, ASTM, ISO 등)을 기재하고, 규격이 없는 경우 자사규격을 기재한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    마. 분량란에는 완제품·부분품 또는 재료 등을 구성하기 위하여 사용되는 원재료·첨가제 또는 색소 등의 분량(단위포함) 및 혼합비를 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    바. 비고란에는 인체접촉여부 및 접촉부위를 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    사. 의약품이 첨가되는 경우에는 해당 의약품의 명칭·성분·규격·분량을 나목 내지 마목에 준하여 기재하고 해당 의약품의 사용목적은 비고란에 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 전기를 사용하는 기구·기계에 해당 하는 경우에는 다음 각 목에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 원재료는 다음 표에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>&nbsp;&nbsp;&nbsp;</CHAR><PICTURE><SHAPEOBJECT InstId="1517942633" ZOrder="8" NumberingType="Figure" TextFlow="BothSides" Lock="false"><SIZE Width="44916" Height="4788" WidthRelTo="Absolute" HeightRelTo="Absolute" Protect="false"/><POSITION TreatAsChar="true" AffectLSpacing="false" VertAlign="Top" HorzAlign="Left" VertOffset="0" HorzOffset="0" FlowWithText="true"/><OUTSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/></SHAPEOBJECT><SHAPECOMPONENT XPos="0" YPos="0" GroupLevel="0" OriWidth="5910" OriHeight="630" HorzFlip="false" VertFlip="false"><ROTATIONINFO Angle="0" CenterX="4960" CenterY="480"/><RENDERINGINFO><TRANSMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><SCAMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/></RENDERINGINFO></SHAPECOMPONENT><IMAGERECT X0="0" Y0="0" X1="5910" Y1="0" X2="5910" Y2="630" X3="0" Y3="630"/><IMAGECLIP Left="0" Top="0" Right="0" Bottom="0"/><INSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/><IMAGE Bright="0" Contrast="0" Effect="RealPic" BinItem="5"/></PICTURE><CHAR> </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 부분품의 명칭란에는 해당 부분품의 일반명칭을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 부분품관리번호란에는 해당 부분품에 대해 모델명 또는 제조회사에서 관리하는 번호 등을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 규격 또는 특성란에는 해당 부분품에 대한 규격이 있는 경우에는 해당 규격(KS, IEC, ISO 등)을 기재하고, 규격이 없는 경우에는 부분품의 기술적 사양(specification)을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    마. 수량란에는 각각의 부분품의 개수를 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    바. 인체에 접촉·삽입되거나 인체에 주입하는 혈액·체액 또는 약물 등에 접촉하거나, 의약품이 첨가되는 의료기기의 경우에는 제1호의 각목의 규정에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    사. 의료기기에 소프트웨어가 사용될 경우에는 가목에 따라 규격 또는 특성란에 소프트웨어의 명칭, 버전을 기재한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 소프트웨어가 단독으로 사용되는 경우에는 다음 표에 따라 해당 란을 각각 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>&nbsp;&nbsp;&nbsp;</CHAR><PICTURE><SHAPEOBJECT InstId="1517942634" ZOrder="8" NumberingType="Figure" TextFlow="BothSides" Lock="false"><SIZE Width="44384" Height="4788" WidthRelTo="Absolute" HeightRelTo="Absolute" Protect="false"/><POSITION TreatAsChar="true" AffectLSpacing="false" VertAlign="Top" HorzAlign="Left" VertOffset="0" HorzOffset="0" FlowWithText="true"/><OUTSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/></SHAPEOBJECT><SHAPECOMPONENT XPos="0" YPos="0" GroupLevel="0" OriWidth="5840" OriHeight="630" HorzFlip="false" VertFlip="false"><ROTATIONINFO Angle="0" CenterX="4960" CenterY="480"/><RENDERINGINFO><TRANSMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><SCAMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/></RENDERINGINFO></SHAPECOMPONENT><IMAGERECT X0="0" Y0="0" X1="5840" Y1="0" X2="5840" Y2="630" X3="0" Y3="630"/><IMAGECLIP Left="0" Top="0" Right="0" Bottom="0"/><INSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/><IMAGE Bright="0" Contrast="0" Effect="RealPic" BinItem="6"/></PICTURE><CHAR> </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 한벌구성의료기기의 경우에는 각각의 의료기기별로 제1호부터 제2호까지의 규정에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제11조(제조방법)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① "제조원의 제조방법에 따른다"라고 기재한다. 다만, 다음 각 호에 해당하는 경우에는 해당사항을 추가하여 기재한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 멸균의료기기의 제조방법의 경우 멸균방법은 별표 2의 멸균방법 또는 이와 동등이상 규격의 멸균방법을 기재한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 최종제품이 동물유래성분을 함유하거나 제조과정 중 동물유래성분을 사용하는 경우 동물의 명칭, 원산국, 연령, 사용부위, 처리공정, 성분명을 기재하고, 그 규격(KS, ASTM, ISO, 등)을 참고하여 적합하게 기재한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 제1항에도 불구하고 제27조에 따른 국제표준화기술문서를 작성하는 경우에는 다음 각 호에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 위탁공정·검사공정 및 멸균공정을 포함하는 원재료 구입부터 최종제품 출하까지의 전체 제조공정에 대한 흐름도를 기재하고, 각 공정에 대한 설명을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제조원 이외의 장소에서 제품설계 및 제조공정을 수행하는 경우 제조공정흐름도에 각 제조소의 제조국, 제조사명, 주소를 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 멸균의료기기의 제조방법의 경우 별표2의 멸균방법 또는 이와 동등이상 규격의 멸균방법 및 멸균조건을 부가하여 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제12조(사용목적)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 사용목적은 다음 각 호에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 사용목적은 근거자료에 따라 적응증, 효능·효과 또는 사용목적을 기재한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제1호의 규정에도 불구하고 시행규칙 제4조에 따른 신고대상 의료기기의 사용목적은 「의료기기 품목 및 품목별 등급에 관한 규정」(식품의약품안전처 고시)에 따라 기재한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. &lt;삭  제&gt; </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 조합의료기기의 경우에는 조합된 기기의 상태로, 한벌구성의료기기의 경우에는 각각의 의료기기별로 제1항의 규정에 따라 사용목적을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 근거가 불명확하거나 막연하고 광범위한 의미의 표현을 하여서는 아니 되며, 중복되거나 지나치게 강조한 표현, 오해 또는 오·남용의 우려가 있는 표현 등을 하여서는 아니 된다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제12조의2(성능)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 성능은 해당 제품이 표방하는 제품의 물리·화학, 전기·기계적 특성, 소프트웨어 특성을 기재한다. 다만, 1등급 의료기기 중 제품의 특성 상 성능을 표방할 수 없는 경우에는 기재하지 아니할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 조합의료기기의 경우에는 조합된 기기의 상태로, 한벌구성의료기기의 경우에는 각각의 의료기기별로 제1항의 규정에 따라 성능을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제13조(사용방법)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 사용방법은 다음 각 호에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 사용방법은 사용 전의 준비사항, 조작방법, 사용 후의 보관 및 관리 방법을 상세히 기재하되, 전문가가 아닌 일반 소비자가 직접 사용하는 의료기기의 경우에는 사용 대상별(장애인, 임산부, 소아 등) 사용에 불편하지 않도록 알기 쉬운 용어로 기재하여야 한다. 다만, 소프트웨어가 단독으로 사용되는 경우에는 사용 전의 준비사항과 사용 후의 보관 및 관리 방법에 대한 기재는 생략할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 사용 전 멸균을 하여야 하는 경우에는 별표 2에 따라 식약처장이 인정하는 멸균방법을 정확히 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 의료기기에 소프트웨어가 내장되거나 단독으로 사용되는 경우에는 프로그램의 기능들을 확인할 수 있는 화면사진과 함께 그 기능에 대한 사용방법을 정확하게 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 일회용 의료기기의 경우 "재사용 금지"를 명확하게 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제14조(사용 시 주의사항)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 사용 시 주의사항은 다음 각 호에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 사용 시 주의사항은 해당 의료기기가 안전하고 합리적으로 사용할 수 있도록 필요한 최신의 안전성 관련 사항을 모두 기재하여야 한다. 이 경우 의학용어사전 등을 참고하여 이해하기 쉽도록 작성하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 다음 각 목에 의한 순서와 요령에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 경고 : 치명적이거나 극히 중대하고 비가역적인 의료기기이상반응이 나타날 경우 또는 의료기기이상반응이 나타난 결과 극히 중대한 사고에 관련될 가능성이 있으므로 특히 주의를 환기할 필요가 있을 경우를 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 의료기기의 특성을 고려한 사용대상 연령, 성별 또는 건강상태 등에 대한 주의사항을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 의료기기의 사용결과 발생할 수 있는 의료기기이상반응 사용상의 부주의에 따른 치명적인 부작용·사고발생 등에 대한 주의사항을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 일반적 주의 : 의료기기로 인한 중대한 사고를 방지하기 위하여 사용 중 주의사항을 기재하고 필요한 경우 사고발생 시 처리방법 등도 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    마. 상호작용 : 다른 의료기기와 병용 시 해당 의료기기가 병용의료기기의 작용을 증강 또는 감약시키거나 의료기기이상반응의 증강이 일어날 경우 또는 새로운 의료기기이상반응이 발생하거나 원질환의 악화 등이 일어날 경우로서 임상적으로 의의가 있는 사항을 기재한다. 다만, 소프트웨어가 단독으로 사용되는 경우에는 이를 생략할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    바. 임부, 수유부, 가임여성, 신생아, 유아, 소아, 고령자에 대한 사용 : 해당 의료기기의 기능적 특성·사용방법 등으로 볼 때 다른 환자에 비하여 특히 주의할 필요가 있다고 판단되는 사항을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    사. 적용상의 주의 : 사용방법 등에 따른 필요한 주의를 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    아. 안전사고의 예방에 필요한 사항이 있는 경우에는 관련 주의사항을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제15조(포장단위)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 의료기기의 포장단위는 제조업자 또는 제조원이 정하는 최소 포장단위로 기재하되, 제조 의료기기의 경우에는 "자사 포장단위"로, 수입 의료기기의 경우에는 "제조원 포장단위"로 기재할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제16조(저장방법 및 사용기간)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 저장방법은 의료기기의 특성을 고려하여 안정성이 보장될 수 있도록 구체적인 보관조건(온도 등) 및 유의사항 등을 병기하여야 한다. 다만, 소프트웨어가 단독으로 사용되는 경우에는 이를 생략할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 사용기간 또는 유효기간은 다음 각 호의 어느 하나에 해당하는 경우에는 식약처장이 고시한 「의료기기의 안정성시험 기준」(식품의약품안전처 고시)에 따라 저장방법 및 사용기간 또는 유효기간을 설정하여 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 멸균의료기기 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 시간이 경과됨에 따라 원재료 등의 물리·화학적 변화로 인한 안전성 또는 성능의 변화가 예측되는 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제17조(시험규격)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 시험규격에는 해당 제품의 안전성 및 성능을 검증하기 위하여 필요한 시험을 다음 각 호에 따라 기재한다. </CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 안전성의 경우 해당 제품의 특성에 따라 식약처장이 고시한 「의료기기의 전기·기계적 안전에 관한 공통 기준규격」, 「의료기기의 전자파 안전에 관한 공통기준규격」, 「의료기기의 생물학적 안전에 관한 공통기준규격」,「의료기기 기준규격」중 해당 규격을 기재하거나 이와 동등한 국제 규격(IEC, ISO 등)을 기재한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 성능은 자사가 설정한 근거에 의한 시험항목, 시험기준 및 시험방법을 다음 각 목에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 시험기준은 시험결과의 적부판정의 기준이 되는 기준치의 허용 범위를 명확히 기재하여야 하며, 시험결과가 온도·습도 등 주위조건에 영향을 받는 경우에는 그 조건을 명시하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 시험방법은 구체적으로 순서에 따라, 시험결과를 정확히 산출할 수 있도록 개조식으로 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 물리·화학적 시험은 인체에 접촉·삽입되거나 인체에 주입하는 혈액·체액 또는 약물 등에 접촉하는 의료기기의 경우 식약처장이 인정한 규격(KS, ISO, ASTM 등)을 기재한다. 다만, 식약처장이 인정한 규격이 없는 제품의 경우 자사가 설정한 근거에 의한 시험항목, 시험기준 및 시험방법을 가목 및 나목에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 조합의료기기의 경우에는 의료기기 전체로서 평가하여야 하는 부분과 각각의 의료기기별로 평가하여야 할 부분의 시험규격을 각각 설정하고, 한벌구성의료기기의 경우에는 각각의 의료기기별로 평가하여야 할 부분의 시험규격을 각각 설정한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제18조(제조원)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 제조원은 다음 각 호에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 수입 의료기기의 경우에는 제조원의 제조국, 제조사명 및 주소를 기재한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 모든 제조공정을 위탁하여 제조하는 경우에는 제조의뢰자와 제조자의 상호와 주소를 모두 기재한다. 다만, 제조자가 외국회사일 경우에는 제조국을 추가로 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제19조(의료기기의 허가·인증·신고의 변경 처리)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 제조(수입)허가·인증을 받거나 신고한 제품의 허가·인증·신고된 항목 중 변경이 있는 경우에는 변경허가·인증을 받거나 변경신고를 하여야 한다. 다만, 다음 각 호의 어느 하나에 해당하는 변경의 경우에는 신규로 허가·인증을 받거나 신고를 하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 작용원리의 변경으로 인한 기 허가·인증·신고사항의 변경</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 해당 품목에 대하여 국내에 최초로 사용하는 원재료의 변경(의료용품 및 체외진단용 의료기기에 한한다)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 별표 4에 따라 안전성이나 유효성에 영향을 미치는 중대한 사항의 변경이 있는 경우에는 시행규칙 제5조제1항제2호의 서류를 제출하여야 한다. 다만, 다음 각 호의 어느 하나에 해당하는 경우에는 시행규칙 제5조제1항제2호의 서류를 제출하지 아니한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 「의료기기법」(이하 "법"이라 한다) 제19조에 따른 기준규격 등의 개정 등으로 식약처장이 변경을 지시한 경우 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 수출만을 목적으로 하는 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 시행규칙 제26조제5항에 따른 경미한 변경 사항은 다음 각 호와 같다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 품목 신고한 제품의 변경(제1항 각 호의 경우에는 제외한다)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 별표 3의 경미한 변경사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ④ 제조업자·수입업자는 제3항에 따른 경미한 변경사항에 대하여는 별지 제1호서식의 경미한 변경사항 보고서(전자문서를 포함한다)를 변경이 있는 날부터 30일 이내 또는 매년 최초 허가·인증 또는 신고일의 전월 말일부터 역산하여 1년 동안의 변경 내용에 대하여 허가·인증 또는 신고일이 속하는 월의 말일까지 식약처장 또는 의료기기정보기술지원센터장에게 제출하여야 한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑤ 제4항에도 불구하고 다음 각 호에 어느 하나에 해당하는 경우에는 변경이 있는 날부터 30일 이내에 별지 제1호서식의 경미한 변경사항 보고서(전자문서를 포함한다)를 식약처장 또는 의료기기정보기술지원센터장에게 제출하여야 한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 생산 및 수입 중단에 따른 일부 모델명의 삭제</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 경미한 변경에 따른 모델명의 변경 또는 추가</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 상호변경에 따른 제조원 명칭변경 및 제품명 변경</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 수입 의료기기의 수출국 제조의뢰자(위탁자를 말함) 및 제조자(수탁자를 말함) 소재지 변경</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. 제조공정 전부를 위탁하는 경우, 제조자 소재지 변경 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑥ 품목류 인증·신고 대상의 경우에는 동일제품군 범위 내에서 자율적으로 변경사항을 관리하도록 하되 추가된 모델명의 목록은 제4항에 따라 제출하여야 한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑦ 제조업자·수입업자는 제3항에 따른 경미한 변경사항은 변경사항이 발생할 때마다 해당 제조(수입)허가증·인증서 뒷면의 변경 및 처분사항 등 란에 변경일자와 변경내용을 기재하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑧ 의료기기 제조업자·수입업자가, 의사의 책임 하에 생리적, 병리적 고유한 특성으로 대체의료기기 또는 치료수단이 없는 환자에게 사용할 목적으로, 담당 의사로부터 다음 각 호의 자료를 구비하여 모양 및 구조의 변경을 요청받아 3D 프린터를 이용하여 제작하는 의료기기(이하 "환자맞춤형 의료기기"라 한다)는 변경허가·인증을 받지 아니하거나 신고하지 아니할 수 있다. 이 경우 허가·인증을 받거나 신고한 제품 당 연간 5회에 한한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 환자맞춤형 의료기기의 적용 타당성에 대한 의사 소견서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 환자맞춤형 의료기기 사용에 대한 환자동의서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 해당분야 의사 5인 이상의 동의서 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑨ 제8항에 따라 허가·인증을 받거나 신고한 모양 및 구조를 변경한 제조업자·수입업자는 해당 의료기기가 사용된 날부터 15일 이내에 별지 제1호의2서식에 따른 환자맞춤형 의료기기 사용 보고서에 제8항 각 호의 서류를 첨부하여 식약처장에게 보고하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제20조(시험용 의료기기 등 확인)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 지방청장은 시행규칙 제32조제2항에 따라 수입업허가 및 수입허가·인증·신고 대상에서 제외하는 의료기기 확인서를 다음 각 호의 구분에 따라 발급한다. 다만, 체외진단용 의료기기를 제외한 제1호에 대한 확인서는 식약처장이 지정한 시험·검사기관장이 발급한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 수입허가·인증을 받기 위한 시험용 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 의료기기 품목지정, 등급심사, 품질관리기준 적합성 평가 및 기술문서 등 심사용 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 제품 개발 등 명확한 연구목적의 연구용 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 제품의 실물 확인 등을 위한 견본용 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. 외국에서 체류하는 동안 사용하던 제품을 귀국 후 계속 사용하거나, 국내에는 허가되지 않은 의료기기(외국 허가제품)를 응급환자 치료에 사용할 목적으로 수입하는 자가 사용용 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  6. 의약품 또는 의료기기 임상시험에 부수적으로 사용되는 별도의 의료기기 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 시험용 의료기기 등을 수입하기 위하여 확인서를 발급받고자 하는 경우에는 별지 제2호서식의 시험용의료기기등 확인서에 다음 각 호의 자료를 첨부하여 지방청장에게 신청하여야 한다. 다만, 체외진단용 의료기기를 제외한 제1항제1호에 대한 확인서는 식약처장이 지정한 시험·검사기관장에게 신청한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 제1항제1호·제2호·제3호·제4호·제6호에 해당하는 경우에는 해당 제품의 모양, 성능, 용도, 수량 등을 확인할 수 있는 자료(사용계획서 포함). 다만, 이미 의료기기 기술문서등 심사를 받은 경우에는 심사결과서만 제출한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제1항제5호에 해당하는 경우에는 해당 제품의 모양, 성능, 용도 등을 확인할 수 있는 자료(사용계획서를 포함한다) 및 신청한 의료기기가 신청인의 치료에 사용하여야 함을 확인할 수 있는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 제1항제6호에 해당하는 경우에는 임상시험계획승인서. 다만, 식약처장의 임상시험계획승인 대상이 아닌 임상시험에 사용되는 의료기기의 경우에는 의료기기 임상시험기관의 승인서를 제출한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 제1항제6호에 해당하는 경우로서 해당 의료기기가 국내에서 허가·인증·신고되어 있지 아니한 경우에는 다음 각 목 중 하나에 해당하는 자료를 추가로 제출한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 수입하고자 하는 의료기기의 제조국 허가 증명자료(제조판매증명서(사본), 허가기관 홈페이지 출력물, CE 표시 등)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 수입하고자 하는 의료기기의 제조국 허가 증명자료를 발급하지 않는 경우에는 기술문서 또는 임상시험자료 제출 대상 심사에 준하는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 제1항에도 불구하고 제1항제6호의 경우에는 식약처장으로부터 승인받은 임상시험 계획승인서에 수입하고자 하는 의료기기에 대한 정보가 없거나, 식약처장의 임상시험 계획승인 대상이 아닌 임상시험에 사용되는 의료기기에 한하여 확인서를 발급한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ④ 지방청장 또는 시험·검사기관장이 시험용의료기기등 확인서를 발급하고자 할 경우에는 제1항 각 호의 어느 하나에 해당하는지와 신청수량(사용계획서 포함)의 타당성 등을 검토하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제20조의2(제조·수입허가 등 의제)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 식약처장이 임상검사실의 품질관리체계 및 검사성능 등을 평가하여 적합하다고 인정하는 임상검사실의 차세대 염기서열분석(Next Generation Sequencing) 검사에 사용하는 유전자서열검사기(A22530.01, 2등급)는 법 제6조에 따라 허가 또는 인증을 받거나 신고한 것으로 본다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제21조(시험·검사의 신청)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 식약처장이 지정한 시험·검사기관에 의료기기 시험·검사를 신청하고자 하는 자는 시험·검사기관에 별지 제4호서식의 신청서와 다음 각 호의 자료를 첨부하여 시험용 의료기기와 함께 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 시행규칙 제5조제1항제2호에 따른 해당 제품의 기술문서 등에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 수입 의료기기의 경우에는 해당 시험용 의료기기를 입증할 수 있는 자료로서 「통합공고」(산업통상자원부 고시)에 따른 표준통관예정보고서, 「수입통관사무처리에 관한 고시」(관세청 고시)에 따른 수입신고필증 등의 사본</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 법 제33조에 따라 전수검사를 받아야 하는 의료기기는 다음 각 호와 같다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 식약처장이 국민건강보호와 안전성·유효성 확보를 위하여 특별히 관리해야 할 필요가 있다고 인정하는 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제22조(중고의료기기 검사필증의 발행 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 의료기기 제조업자 또는 수입업자가 의료기관으로부터 자기 회사가 제조 또는 수입한 의료기기를 구입한 경우에는 시행규칙 제27조제1항제15호 또는 제33조제1항제19호에 따른 검사를 하고, 적합한 경우에 별지 제10호서식의 검사필증을 해당 의료기기의 외장에 부착하여 출고하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 판매업자 또는 임대업자는 의료기관으로부터 의료기기를 구입한 경우에는 시행규칙 제39조제1호에 따라 해당 의료기기의 제조업자, 수입업자 또는 의료기기 시험·검사기관에 별지 제11호서식의 신청서를 첨부하여 검사를 의뢰한 후 검사필증이 부착된 의료기기에 한하여 판매 또는 임대하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 제조업자 또는 수입업자는 제2항에 따라 판매업자 또는 임대업자로부터 검사를 의뢰받은 경우에는 의뢰를 받은 날부터 30일 이내에 시행규칙 제27조제1항제16호 또는 제33조제1항제20호에 따른 검사를 실시하고 의뢰인에게 별지 제12호서식에 따라 의뢰인에게 그 결과를 통보하고, 적합한 경우에는 별지 제10호서식의 검사필증을 발행하여야 하며, 해당 의료기기의 외장에 검사필증을 부착하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ④ 제3항에도 불구하고 제조업자 또는 수입업자가 30일 이내에 검사필증 발행이 어려운 경우에는 그 지연사유 및 처리기한을 기재하여 해당 판매업자 또는 임대업자에게 회신하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑤ 제1항 및 제2항에도 불구하고 시행규칙 제41조에 따라 다음 각 호의 어느 하나에 해당하는 의료기기는 출고 또는 판매·임대할 때 중고의료기기 검사필증을 부착하지 아니할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 「의료법」 제37조에 따른 진단용방사선발생장치</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 1등급 의료기기. 다만, 다음 각 목에 해당하는 의료기기는 제외한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 전기를 사용하는 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 방사선, 레이저 등 방어용 기구  </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. &lt;삭  제&gt; </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 인체에 에너지를 전달하는 기기  </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑥ 수입업자가 허가·인증 또는 신고된 의료기기를 중고로 수입하는 경우에는 제1항을 준용한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제4장 기술문서 등 심사</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제23조(심사대상 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 기술문서 등의 심사는 제조·수입허가·인증(변경허가·인증을 포함한다)을 받고자 하는 의료기기를 대상으로 한다. </CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 2등급 의료기기의 경우에는 별지 제3호서식의 본질적 동등품목 비교표를 사용하여 별표 5의 동등제품 판단기준에 따라 새로운제품, 개량제품, 동등제품으로 구분하여 심사한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 제1항에도 불구하고 다음 각 호에 해당하는 의료기기는 기술문서 등의 심사대상에서 제외한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 수출용의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 동등공고제품</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 제3조제11항에 따라 동일제품임을 확인받은 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 경미한 변경 대상 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제24조(신속심사 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 식약처장은 다음 각 호의 의료기기에 대하여 우선적으로 신속하게 심사하여 허가할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 첨단의료기기 및 희소의료기기 등 산업발전 또는 환자 치료를 위하여 필요한 경우 식약처장이 지정한 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 이미 허가(신고)한 체외진단용 의료기기로 측정이 불가능하여 신속한 도입이 필요하다고 판단되어 식약처장이 지정한 체외진단용 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제25조(심사기준 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 식약처장 또는 기술문서심사기관의 장은 시행규칙 제9조제2항에 따른 의료기기 기술문서 등 심사의뢰서(이하 "심사의뢰서"라 한다)를 제출받은 경우에는 해당 제품의 제조업자(수입업자)별, 품목류별 또는 품목별(모델명)로 심사하는 것을 원칙으로 하며, 심사의뢰서를 제출받은 후 추가된 제품은 심사에서 제외한다. 다만, 동일제품군에 대하여는 하나의 기술문서로 심사할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 심사의뢰서에서 사용하는 용어 및 기호 등은 한국산업규격(KS), 국제규격(IEC, ISO 등), 대한약전(KP) 및 식약처장이 인정하는 공정서(USP, JP 등)를 따른다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제26조(심사자료의 종류 및 범위 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 기술문서 등의 심사를 위하여 제출하여야 하는 자료의 종류는 다음 각 호와 같다. </CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 이미 허가받은 제품과 비교한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 사용목적에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 작용원리에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 제품의 성능 및 안전을 확인하기 위한 다음 각 목의 자료로서 시험규격 및 그 설정근거와 실측치에 관한 자료. 다만, 국내 또는 국외에 시험규격이 없는 경우에는 기술문서 등의 심사를 받으려는 자가 제품의 성능 및 안전을 확인하기 위하여 설정한 시험규격 및 그 근거와 실측치에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 전기·기계적 안전에 관한 자료(전기·전자회로를 사용하는 기구·기계·장치에 한한다)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 생물학적 안전에 관한 자료(인체에 직·간접적으로 접촉하여 생물학적 안전에 대한 확인이 필요한 의료기기에 한한다)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    1) 독성시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     가) 세포독성시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     나) 전신독성(급성)시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     다) 아만성(아급성) 독성시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     라) 만성독성시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     마) 유전독성시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     바) 생식독성시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     사) 독성동태시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     아) 면역독성시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    2) 자극성과 감작성 시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    3) 이식시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    4) 혈액적합성 시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    5) 발암성 시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    6) 생분해성 시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    7) 에틸렌옥사이드 잔류량 시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    8) 무균시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 방사선에 관한 안전성 자료(방사선을 이용하거나 방사선에 노출되는 의료기기에 한한다)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 전자파안전에 관한 자료(전기·전자 회로를 사용하는 의료기기에 한한다)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    마. 성능에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    바. 물리·화학적 특성에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    사. 안정성에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. 기원 또는 발견 및 개발경위에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  6. 임상시험에 관한 자료 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  7. 외국의 사용현황 등에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 기술문서 등 심사를 위하여 제출하는 제1항제4호나목의 생물학적 안전에 관한 자료의 범위는 별표 12와 같다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제27조(국제표준화기술문서 작성)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 제26조에도 불구하고 4등급 의료기기의 경우에는 국제표준화기술문서로 작성하여야 한다. 다만, 체외진단용 의료기기의 경우에는 제외한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 4등급 의료기기 이외의 품목은 신청인이 원하는 경우 국제표준화 기술문서로 작성할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 국제표준화기술문서의 세부 작성 방법은 별표 10에 따른다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ④ 국제표준화기술문서의 제출자료의 종류는 별표 11의 비교에 따라 제26조에서 정한 심사자료의 종류로 보고 제28조(심사자료의 면제), 제29조(첨부자료의 요건), 제30조(자료의 작성 등), 제31조(기술문서 심사결과통지서 변경)를 적용한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제28조(심사자료의 면제)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 제23조제2항에 따라 새로운제품, 개량제품 및 동등제품으로 구분된 2등급 의료기기의 경우에는 별표 7에 따라 다음 각 호에 해당하는 자료의 제출을 면제할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 제26조제2호·제3호 및 제5호부터 제7호까지의 규정에 따른 자료 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제26조제4호 각 목의 규정에 따른 자료에 한하여 시험성적서를 제외한 시험규격의 설정근거와 실측치자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 3·4등급 의료기기의 경우에는 제26조제5호부터 제7호까지의 규정에 따른 자료에 한하여 별표 7에 따라 제출을 면제할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 식약처장이 인정하는 원재료 또는 완제품에 관한 규격(KS, ISO, ASTM)에 적합한 의료기기의 경우에는 제26조제4호나목의 규정에 따른 자료를 제출하지 아니할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ④ 제26조의 규정에도 불구하고 시험자체가 이론적·기술적으로 실시 불가능하거나 실시 가능하더라도 실시하는 것이 무의미하다고 인정되는 경우에는 해당 자료의 제출을 면제할 수 있다.(예: 실버설파다이아진 크림이 함유된 창상피복재의 경우 각 원재료의 안전성이 확보 되었으나, 실버설파다이아진 크림의 항균작용으로 인하여 세포독성 시험결과가 부적합이 나올 수밖에 없어 해당 시험이 무의미한 경우 등)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제29조(첨부자료의 요건)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 기술문서 등의 심사를 위한 첨부자료의 요건은 다음 각 호와 같다. 다만, 제26조제4호의 규정에 따른 시험자료의 경우에는 제출일을 기준으로 발급일로부터 3년이 경과된 시험자료는 해당 제품이 시험 이후에 변경이 없음을 확인하는 자료를 추가로 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 이미 허가·인증받은 제품과 비교한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    기존 허가·인증받은 제품과 명칭(제품명, 품목명, 모델명), 제조(수입)업소명, 제조원 및 소재지, 허가(인증)번호, 사용목적, 작용원리, 원재료, 성능, 시험규격, 사용방법 등을 비교한 별지 제3호서식의 비교표</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 사용목적에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    해당 제품의 적응증, 사용목적(효능·효과)에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 작용원리에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    해당 제품의 사용목적을 달성하기 위해 영향을 미치는 물리·화학·전기·기계적 작용원리에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 전기·기계적 안전에 관한 자료 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 일반사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      전기를 사용하는 의료기기의 경우에 제출하여야 하며, 다음 중 어느 하나에 해당되는 자료로서 해당 제품과 모델명이 동일하여야 한다. 다만, 개발 시 명칭 등으로 자료상의 모델명과 해당 제품의 모델명이 동일하지 않은 경우에는 이를 입증하는 자료를 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      1) 식약처장이 지정한 시험·검사기관에서 발급한 시험성적서 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      2) 국제전기기술위원회(IEC)가 운영하는 국제전기기기인증제도(IECEE CB-Scheme)에 따라 국제공인시험기관(NCB : National Certification Body)에서 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      3) 한국인정기구(KOLAS: Korea Laboratory Accreditation Scheme)(이하 "KOLAS"라 한다)에서 인정한 의료기기 분야의 시험검사기관에서 인정된 규격코드로 적합하게 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      4) 해당 의료기기에 대하여 경제협력개발기구(OECD) 회원국에 허가 당시 제출되어 평가된 시험성적서로 해당 정부 또는 정부가 허가 업무를 위임한 등록기관이 제출받아 승인하였음을 확인한 자료 또는 이를 공증한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 기준 및 시험방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      「의료기기의 전기·기계적 안전에 관한 공통기준규격」(식품의약품안전처 고시) 및 「의료기기 기준규격」(식품의약품안전처 고시)을 따르거나 이와 동등 이상의 국제규격(IEC 등)에 따르되, 「전기사업법」에 의한 국내 표준전압, 표준주파수, 표준전원플러그 등의 기준을 따라야 한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. 생물학적 안전에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 일반사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      인체에 접촉·삽입되거나 인체에 주입하는 혈액·체액 또는 약물 등과 접촉하는 의료기기 또는 부분품의 경우 제출하여야 하며, 다음 중 어느 하나에 해당하는 자료로서 해당 제품과 모델명이 동일하여야 한다. 다만, 개발 시 명칭 등으로 자료상의 모델명과 해당 제품의 모델명이 동일하지 않은 경우에는 이를 입증하는 자료를 제출하여야 하며, 무균시험 및 EO가스 잔류량 시험의 경우에는 「의료기기 제조 및 품질관리기준」(식품의약품안전처 고시) 또는 이와 동등 이상의 규격에 따른 제조사의 품질관리시스템 하에서 실시한 시험성적서 또는 적합함을 입증하는 자료를 제출할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    1) 식약처장이 지정한 시험·검사기관에서 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    2) 경제협력개발기구(OECD)로부터 비임상관리기준(GLP)을 준수하는 OECD 회원국 또는 이를 준수하는 것으로 OECD로부터 인정받은 비회원국의 비임상시험 실시기관에서 발급한 시험자료 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    3) 1) 또는 2)에 해당하는 자료로서 해당 제품과 원재료가 동일하고 인체접촉시간·인체접촉부위 등이 동등하거나 동등이상인 제품의 생물학적 안전에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 기준 및 시험방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    「의료기기의 생물학적 안전에 관한 공통기준규격」(식품의약품안전처 고시) 및 「의료기기 기준규격」(식품의약품안전처 고시)을 따르거나 이와 동등 이상의 국제규격(ISO 등)에 따른다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  6. 방사선에 관한 안전성 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 일반사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    방사선을 이용하는 의료기기이거나 방사선에 노출되는 등 해당 의료기기가 방사선에 관한 안전성이 요구되는 경우 제출하여야 하며, 다음 중 어느 하나에 해당되는 자료로서 해당 제품과 모델명이 동일하여야 한다. 다만, 개발 시 명칭 등으로 자료상의 모델명과 해당 제품의 모델명이 동일하지 않은 경우에는 이를 입증하는 자료를 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    1) 식약처장이 지정한 시험·검사기관에서 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    2) 국제전기기술위원회(IEC)가 운영하는 국제전기기기인증제도(IECEE CB-Scheme)에 따라 국제공인시험기관(NCB : National Certification Body) 등에서 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    3) KOLAS에서 인정한 의료기기분야 시험검사기관에서 인정된 규격코드로 적합하게 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    4) 해당 의료기기에 대하여 경제협력개발기구(OECD) 회원국에 허가 당시 제출되어 평가된 시험성적서로 해당정부 또는 정부가 허가 업무를 위임한 등록기관이 제출받아 승인하였음을 확인한 자료 또는 이를 공증한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 기준 및 시험방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      「의료기기의 전기·기계적 안전에 관한 공통기준규격」(식품의약품안전처 고시)[별표 2] 의료기기의 방사선 안전에 관한 보조기준규격, 「의료기기 기준규격」(식품의약품안전처 고시) 또는 이와 동등이상의 국제규격(IEC 등)에 따르되, 「전기사업법」에 의한 국내 표준전압, 표준주파수, 표준전원플러그 등의 기준을 따라야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  7. 전자파안전에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 일반사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      전자파 안전성이 요구되는 의료기기 또는 부분품의 경우 제출하여야 하며, 다음 중 어느 하나에 해당되는 자료로서 해당 제품과 모델명이 동일하여야 한다. 다만, 개발 시 명칭 등으로 자료상의 모델명과 해당 제품의 모델명이 동일하지 않은 경우에는 이를 입증하는 자료를 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    1) 식약처장이 지정한 시험·검사기관에서 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    2) 국제전기기술위원회(IEC)가 운영하는 국제전기기기인증제도(IECEE CB-Scheme)에 따라 국제공인시험기관(NCB : National Certification Body)에서 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    3) KOLAS에서 인정한 의료기기 분야 시험검사기관에서 인정된 규격코드로 적합하게 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    4) 해당 의료기기에 대하여 경제협력개발기구(OECD) 회원국에 허가 당시 제출되어 평가된 시험성적서로 해당정부 또는 정부가 허가 업무를 위임한 등록기관이 제출받아 승인하였음을 확인한 자료 또는 이를 공증한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 기준 및 시험방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      「의료기기의 전자파안전에 관한 공통기준규격」(식품의약품안전처 고시) 및 「의료기기 기준규격」(식품의약품안전처 고시)을 따르거나 이와 동등이상의 국제규격(IEC 등)에 따르되, 「전기사업법」에 의한 국내 표준전압, 표준주파수, 표준전원플러그 등의 기준을 따라야 한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  8. 성능에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 일반사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      다음 중 어느 하나에 해당되어야 하며, 해당 제품과 모델명이 동일하여야 한다. 다만, 개발 시 명칭 등으로 자료상의 모델명과 해당 제품의 모델명이 동일하지 않은 경우에는 이를 입증하는 자료를 제출하여야 하며, 소프트웨어가 내장되어 있거나 단독으로 사용되는 경우에는 별표 13에 따른 별지 제13호서식의 적합성 확인보고서와 소프트웨어 검증 및 유효성 확인 자료를 제출하여야 하고, 동물을 대상으로 한 성능 확인이 필요한 경우 동물시험 자료를 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      1) 식약처장이 지정한 시험·검사기관에서 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      2) 대학 또는 연구기관 등 국내·외의 전문기관에서 시험한 것으로서 해당 기관의 장이 발급하고 그 내용(기관의 시험시설 개요, 주요설비, 시험자의 연구경력 등을 포함한다)을 검토하여 타당하다고 인정할 수 있는 시험성적서. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      3) 「의료기기 제조 및 품질관리기준」 또는 이와 동등 이상의 규격에 따른 제조사의 품질관리시스템 하에서 실시한 제품의 성능에 관한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 기준 및 시험방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    자사의 기준 및 시험방법에 따른다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  9. 물리·화학적 특성에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 일반사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      인체에 접촉·삽입되거나 인체에 주입하는 혈액·체액 또는 약물 등에 접촉하는 의료기기의 경우 해당되는 부분에 대한 화학구조, 적외선흡수, 자외선흡수, 원자흡광도, 융점, 비점, 내구성, 경도, 색조, 용출물, 표면특성 자료, 동물유래성분을 사용하는 경우 해당 규격(KS, ASTM, ISO 등)에 따른 동물의 명칭, 원산국, 연령, 사용부위, 처리공정, 성분명에 대한 자료 등은 다음 각 단의 어느 하나에 해당되어야 하며, 해당 제품과 모델명이 동일하여야 한다. 다만, 개발 시 명칭 등으로 자료상의 모델명과 해당 제품의 모델명이 동일하지 않은 경우에는 이를 입증하는 자료를 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      1) 식약처장이 지정한 시험·검사기관에서 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      2) 대학 또는 연구기관 등 국내·외의 전문기관에서 시험한 것으로서 해당 전문기관의 장이 발급하고 그 내용(기관의 시험시설 개요, 주요설비, 시험자의 연구경력 등을 포함한다)을 검토하여 타당하다고 인정할 수 있는 시험성적서 또는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      3) 「의료기기 제조 및 품질관리기준」 또는 이와 동등 이상의 규격에 따른 제조사의 품질관리시스템 하에서 실시한 물리·화학적 특성에 관한 시험성적서 또는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 기준 및 시험방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      식약처장이 인정한 원재료 및 완제품에 대한 규격에 적합한 기준 및 시험방법에 따른다. 다만, 식약처장이 인정하는 규격이 없는 경우에는 자사의 기준 및 시험방법에 따른다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  10. 안정성에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 일반사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      다음 중 어느 하나에 해당되는 자료로서, 해당 제품과 모델명이 동일하여야 한다. 다만, 개발 시 명칭 등으로 자료상의 모델명과 해당 제품의 모델명이 동일하지 않은 경우에는 이를 입증하는 자료를 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      1) 식약처장이 지정한 시험·검사기관에서 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      2) 대학 또는 연구기관 등 국내·외의 전문기관에서 시험한 것으로서 해당 전문기관의 장이 발급하고 그 내용(전문기관의 시험시설 개요, 주요설비,  시험자의 연구경력 등을 포함한다)을 검토하여 타당하다고 인정할 수 있는 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      3) 「의료기기 제조 및 품질관리기준」 또는 이와 동등 이상의 규격에 따른 제조사의 품질관리시스템 하에서 실시한 제품의 안정성에 관한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 기준 및 시험방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     「의료기기의 안정성시험 기준」에 따른다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  11. 기원 또는 발견 및 개발경위에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     해당 제품에 대해 심사에 도움을 줄 수 있도록 육하원칙(예: 언제, 어디서, 누가, 새로 발견한 작용원리 및 원재료, 기초시험·임상시험 등에 들어간 것은 언제, 어디서 였나 등)에 따라 명료하게 기재된 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  12. 임상시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 일반사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      의료기기 허가를 위한 임상시험에 사용되는 의료기기의 안전성 및 유효성을 증명하기 위하여 사람을 대상으로 시험한 자료로서 다음 중 어느 하나에 해당되어야 한다. 이 경우 1, 2등급 의료기기의 경우에는 신청한 제품과 동등한 제품의 임상시험에 관한 자료(논문, 문헌 등)를 제출할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      1) 식약처장이 지정한 임상시험기관에서 시험한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    2) 외국자료로서 그 내용을 검토하여 실시기관의 신뢰성이 인정되고 「의료기기 임상시험 관리기준」(의료기기법 시행규칙 별표2의2)에 의하여 실시한 것으로 판단되는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    3) 해당 의료기기에 대하여 경제협력개발기구(OECD) 회원국에 허가 당시 제출되어 평가된 임상시험에 관한 자료로서 해당 정부 또는 정부가 허가 업무를 위임한 등록기관이 제출받아 승인하였음을 확인한 자료 또는 이를 공증한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    4) 과학논문인용색인(Science Citation Index)에 등재된 전문학회지에 게재된 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 가목의 해당하는 자료는 다음의 사항을 포함하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      1) 임상시험방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>       임상시험방법은 다음의 사항을 포함해야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     가) 피험자의 선정기준, 제외기준 및 목표한 피험자의 수</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="40" Style="0"><TEXT CharShape="5"><CHAR>        원칙적으로 하나의 적응증마다 해당 의료기기의 특성과 임상시험방법 등을 종합적으로 고려하여, 통계적으로 타당하게 임상시험 예수가 결정되었음을 입증하는 자료. 다만, 적응질환의 발생 증례 자체가 적어 임상시험 예수의 확보가 현실적으로 곤란한 경우에는 이를 입증할 수 있는 자료를 추가로 첨부하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     나) 조작방법 또는 사용방법과 그 설정사유</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     다) 비교시험용 의료기기를 사용하는 경우 그 선택사유</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     라) 병용사용의 유무</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     마) 관찰항목, 측정항목, 임상검사항목, 측정기준 및 검사방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     바) 유효성 평가기준, 평가방법 및 해석방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     사) 부작용을 포함한 안전성의 평가기준 및 시험방법</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    2) 임상결과</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      임상결과는 다음의 사항을 포함해야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     가) 임상시험의 성적(임상례에 대한 계획된 수, 실제 대상수, 완료된 수, 중도탈락자 수 및 이유 등을 포함하며, 이 경우 피험자별 부작용 등에 대한 사항이 포함되어야 한다)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     나) 증례기록 요약</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     다) 기타 임상시험성적의 확인에 필요한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    3) 임상평가</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>       해당 적응증에 대한 의료기기의 유효율이 의학적·한의학적 원리에 기준하여 임상적 유의성이 있음을 입증하는 자료로 그 타당성이 판단되는 경우 이를 인정할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 기타</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      1) 민족적 요인의 차이가 있어 외국임상시험에 관한 자료를 그대로 적용하기가 어렵다고 판단되는 경우, 식약처장은 국내에서 우리나라 사람을 대상으로 한 자료를 추가 제출할 것을 요구할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="39" Style="0"><TEXT CharShape="5"><CHAR>      2) 생명을 위협하는 희귀한 질환에 적용되는 희소의료기기의 경우에는 식약처장이 타당하다고 인정하는 범위의 임상시험에 관한 자료로 갈음할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  13. 외국의 사용현황 등에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     해당 의료기기 유효성에 대한 심사에 도움을 줄 수 있도록 각 국가의 사용현황에 관한 자료로서 외국의 판매 또는 허가현황, 사용 시 보고된 부작용, 제조허가 경위 등과 관련된 자료, 제조국에서 사용되지 않는 경우는 그 사유</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제30조(자료의 작성 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 의료기기 기술문서 등의 심사를 받고자 하는 자는 시행규칙 별지 제8호서식에 따라 작성하여야 하고, 제29조 또는 제33조에 따른 첨부자료(전자문서를 포함한다) 등을 식약처장 또는 기술문서심사기관의 장에게 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 제출자료는 제29조 또는 제33조에 따른 요건에 적합하여야 한다. 다만, 해당 제품의 특성상 첨부자료의 일부가 불필요한 경우에는 그 사유를 구체적으로 기재하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 외국의 자료는 주요사항을 발췌한 한글 요약문 및 원문을 첨부하여야 하며, 필요한 경우에 한하여 번역물을 요구할 수 있다. 다만, 영어 외의 외국어 자료는 공증된 전체 번역문을 첨부하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제31조(기술문서 심사결과통지서 변경)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 시행규칙 제9조제4항에 따른 의료기기 기술문서 등의 심사결과통지서를 변경하고자 하는 자는 시행규칙 별지 제8호서식에 따른 심사의뢰서를 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 제1항의 규정에 의한 변경신청 시 첨부하는 서류는 다음 각 호와 같다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 변경대비표</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제26조 또는 제33조에서 규정하는 자료 중 변경사항에 대한 근거자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 제2항제1호의 서류는 다음 표에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>&nbsp;&nbsp;&nbsp;</CHAR><PICTURE><SHAPEOBJECT InstId="1517942635" ZOrder="8" NumberingType="Figure" TextFlow="BothSides" Lock="false"><SIZE Width="44688" Height="4180" WidthRelTo="Absolute" HeightRelTo="Absolute" Protect="false"/><POSITION TreatAsChar="true" AffectLSpacing="false" VertAlign="Top" HorzAlign="Left" VertOffset="0" HorzOffset="0" FlowWithText="true"/><OUTSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/></SHAPEOBJECT><SHAPECOMPONENT XPos="0" YPos="0" GroupLevel="0" OriWidth="5880" OriHeight="550" HorzFlip="false" VertFlip="false"><ROTATIONINFO Angle="0" CenterX="4960" CenterY="480"/><RENDERINGINFO><TRANSMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><SCAMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/></RENDERINGINFO></SHAPECOMPONENT><IMAGERECT X0="0" Y0="0" X1="5880" Y1="0" X2="5880" Y2="550" X3="0" Y3="550"/><IMAGECLIP Left="0" Top="0" Right="0" Bottom="0"/><INSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/><IMAGE Bright="0" Contrast="0" Effect="RealPic" BinItem="7"/></PICTURE><CHAR> </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제5장 체외진단용 의료기기의 기술문서 등 심사 등</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제32조(체외진단용 의료기기의 허가·인증신청서 및 신고서 항목 기재요령 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 제9조·제10조, 제12조부터 제14조, 제16조·제17조의 규정에도 불구하고 체외진단용 의료기기의 허가·인증신청서 및 신고서 항목은 다음 각 호에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 모양 및 구조는 해당 제품의 개요를 포함하여 모양·구조·중량 및 치수 등을 기재하며, 액상 또는 분말인 경우에는 색, 성상, 액성, 냄새 등 외관상 특징을 포함하여 기재한다. 이 경우 제품의 개요는 해당 제품을 개발하기 위하여 적용한 원리 등을 포함하여 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 원재료는 다음 각 목에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 다음 표의 내용에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>&nbsp;&nbsp;&nbsp;</CHAR><PICTURE><SHAPEOBJECT InstId="1517942636" ZOrder="8" NumberingType="Figure" TextFlow="BothSides" Lock="false"><SIZE Width="50996" Height="5548" WidthRelTo="Absolute" HeightRelTo="Absolute" Protect="false"/><POSITION TreatAsChar="true" AffectLSpacing="false" VertAlign="Top" HorzAlign="Left" VertOffset="0" HorzOffset="0" FlowWithText="true"/><OUTSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/></SHAPEOBJECT><SHAPECOMPONENT XPos="0" YPos="0" GroupLevel="0" OriWidth="6710" OriHeight="730" HorzFlip="false" VertFlip="false"><ROTATIONINFO Angle="0" CenterX="4960" CenterY="480"/><RENDERINGINFO><TRANSMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><SCAMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/><ROTMATRIX E1="1.00" E2="0.00" E3="0.00" E4="0.00" E5="1.00" E6="0.00"/></RENDERINGINFO></SHAPECOMPONENT><IMAGERECT X0="0" Y0="0" X1="6710" Y1="0" X2="6710" Y2="730" X3="0" Y3="730"/><IMAGECLIP Left="0" Top="0" Right="0" Bottom="0"/><INSIDEMARGIN Left="0" Right="0" Top="0" Bottom="0"/><IMAGE Bright="0" Contrast="0" Effect="RealPic" BinItem="8"/></PICTURE><CHAR> </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 명칭란에는 해당 구성제품의 일반명칭을 기재한다. 키트 또는 세트로 되어있는 경우에는 구성시약별로 구분하여 기재하며, 두 세트 이상이 함께 사용되어 하나의 사용목적을 달성하는 경우에는 세트별로 구분하여 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 배합목적과 원재료명 또는 성분명란에는 체외진단용 의료기기의 특성에 맞게 배합하는 목적과 각 구성시약의 일반명 또는 화학명을 각각 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 분량란에는 각 성분의 분량(역가, 소요량 등) 및 단위(mL, mg, v/v, w/v, w/w 등)를 기재하고 범위를 설정할 수 있다. 다만, 주성분 이외의 성분의 경우에는 "적량"으로 표시할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    마. 규격란에는 원재료에 대한 규격이 있는 경우에는 해당 규격(KP, USP 등)을 기재하고, 규격이 없는 경우에는 자사규격 등을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    바. 비고란에는 각 구성 시약의 총량 및 수량 등을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 사용목적은 다음 각 목에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 사용목적은 근거자료에 따라 검사대상, 검체종류, 검사항목, 측정원리 및 결과판정방법(정성 또는 정량 등) 등을 구체적으로 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 성능은 체외진단용 의료기기의 특성에 맞는 성능을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 하나의 시약이 두 개 이상의 사용목적 및 성능을 갖는 경우에는 각각의 사용목적과 성능을 모두 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 사용방법은 검체준비 및 저장방법, 검사 전 준비사항, 검사과정, 결과판정 및 정도관리 등이 포함되도록 기재한다. 다만, 사용된 시약의 양은 성능이 확인될 수 있는 구체적 분량(범위)을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. 사용 시 주의사항은 다음 각 목의 사항을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 체외진단용으로 사용해야 함</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 일반적인 실험실 안전 및 생물학적 위험물질(검체, 감염 가능성 물질 및 폐기물 등) 취급 시 안전 등의 주의사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 경고사항, 제품 취급 및 보관상의 주의, 적용상의 주의, 결과 판정상 주의 및 제품 폐기 시 주의 등에 관한 내용</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 다른 의료기기와 결합하여 사용하는 경우에는 조합에 대한 정보</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  6. 저장방법 및 사용기간은「의료기기의 안정성시험 기준」(식품의약품안전처 고시)에 따라 저장방법 및 사용기간 또는 유효기간을 설정하여 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 키트 또는 세트 제품인 경우에는 구성품별 보관온도 등을 각각 기재하고 사용기간 또는 유효기간이 각각 다른 경우에는 가장 짧은 제품의 기간을 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 개봉 후 시약의 저장방법 및 사용기간 또는 유효기간 등이 포함되도록 기재한다. 다만, 일회용으로서 낱개 포장 되어있는 체외진단용 의료기기는 제외한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  7. 시험규격은 해당 체외진단용 의료기기의 품질관리에 적정을 기할 수 있도록 제품의 특성에 따라 성능 등을 고려하여 제조단위별로 적용할 수 있는 시험규격을 제17조제1항제2호가목·나목에 따라 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 조합의료기기 및 한벌구성의료기기의 경우에는 제8조부터 제18조까지의 해당 규정을 준용하여 기재한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제33조(체외진단용 의료기기의 심사자료의 종류 및 요건)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 제26조에도 불구하고 체외진단용 의료기기의 경우에는 다음 각 호의 자료를 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 개발경위, 측정 원리·방법 및 국내외 사용현황에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 원재료 및 제조방법에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 사용목적에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 저장방법과 사용기간 또는 유효기간에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. 제품의 성능을 확인하기 위한 다음 각 목의 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 분석적 성능시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 임상적 성능시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 품질관리 시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 표준물질 및 검체보관 등에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  6. 체외진단용 의료기기의 취급자 안전에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  7. 이미 허가·인증받은 제품과 비교한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 제29조에도 불구하고 제1항에 따른 체외진단용 의료기기의 기술문서 등의 심사를 위한 첨부자료의 요건과 근거자료는 다음 각 호와 같다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 개발경위, 측정 원리·방법 및 국내외 사용현황에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 개발경위는 측정하고자 하는 대상 또는 질병이나 증후군의 설명과 개발배경이 포함된 논문, 문헌 등 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 측정원리 및 방법은 해당제품의 측정 및 질병진단 목적을 달성하기 위하여 적용된 원리에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 국내외 사용현황에 관한 자료는 다음의 사항을 포함한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    1) 국내외의 판매 또는 허가현황 및 제조허가(인증) 경위 등과 관련된 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    2) 사용시 보고된 측정오류</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    3) 제조국에서 사용되지 않는 경우에는 그 사유</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 원재료 및 제조방법에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    원재료의 성분 및 분량 등을 확인할 수 있는 근거자료와 제조공정의 흐름도를 포함한 제조공정에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 사용목적에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    해당 제품의 검사대상, 검체종류, 검사항목, 측정원리 및 결과판정방법(정성, 정량 등)에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 저장방법과 사용기간 또는 유효기간에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    완제품 및 개봉 후 시약의 안정성에 관한 자료로서 식약처장이 정한 기준에 따라 설정된 안정성에 관한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. 제품의 성능을 확인하기 위한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 제품의 성능을 확인하기 위한 자료는 다음의 자료를 포함한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    1) 분석적 성능시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     분석적 성능시험자료에는 다음의 평가항목을 포함한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     가) 분석적민감도(판정기준치(cut-off value), 최소검출한계, 측정범위 등)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     나) 분석적특이도(교차반응 등)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     다) 정밀도(반복, 재현성 등)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     라) 정확도</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    2) 임상적 성능시험에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    체외진단용 의료기기의 성능 및 유효성을 입증하기 위하여 사람에서 유래된 검체를 대상으로 시험한 자료로서 다음의 평가항목을 포함한다. 다만, 민족적 요인의 차이가 있어 외국 임상적 성능 시험을 그대로 적용하기가 어렵다고 판단되는 경우, 식약처장은 국내에 거주하는 한국인으로부터 유래한 검체를 대상으로 한 자료를 추가 제출할 것을 요구 할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     가) 임상적민감도</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     나) 임상적특이도</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    3) 완제품의 품질관리 시험성적서 또는 품질관리 시험에 관한 자료(3배치 1회 이상 또는 1배치 3회 이상)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    4) 표준물질 및 검체보관 등에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     가) 완제품 품질관리 시험에 사용된 표준물질에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>     나) 검체보관 및 취급상(온도, 습도 등)의 조건 설정 근거 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 가목의 1), 2)에 해당하는 자료는 국내·외 허가·인증받은 체외진단용 의료기기와의 상관성을 확인할 수 있는 비교시험성적서를 포함하여야 한다. 다만, 측정원리 및 측정항목이 새로운 경우에는 동일목적으로 사용되는 제품과 비교할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 다음 어느 하나에 해당하는 시험의 경우에는 법 제10조에 따라 실시한 가목의 2)에 해당하는 자료를 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    1) 4등급 체외진단용 의료기기의 시험</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    2) 인체로부터 검체를 채취하는 방법의 위해도가 큰 시험(검체의 채취방법이 인체의 피부, 점막, 안구, 요도를 침투 또는 관통하거나, 외이도, 외비공, 인두, 직장 또는 자궁경부를 넘어서 귀, 코, 입, 항문관 또는 질에 들어가는 침습적인 시험. 다만, 정맥채혈 등 피험자에게 중대한 위험을 미치지 않는 시험 및 잔여검체로 실시하는 시험은 제외)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    3) 이미 확립된 의학적 진단방법 또는 허가·인증받은 체외진단용 의료기기로 임상적 성능시험의 결과를 확인할 수 없는 시험</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    4) 의약품 등과 함께 동반하여 진단하는 시험(다만, 이미 허가·인증받은 의료기기와 사용목적, 작용원리 등이 동등하지 아니한 체외진단용 의료기기의 시험에 한함)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  6. 체외진단용 의료기기의 취급자 안전에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 인간혈액 유래물질이 포함되었을 경우 사람면역결핍바이러스(HIV), C형간염바이러스(HCV), B형간염바이러스(HBV)가 음성 또는 불활성화하여 감염력이 없음을 입증하는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 유해물질(독성, 가연성 등) 등 취급자 안전 및 적합성을 확인한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  7. 이미 허가·인증받은 제품과 비교한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    이미 허가·인증받은 제품과 명칭(제품명, 품목명, 모델명), 제조(수입)업소명, 제조원 및 소재지, 허가(인증)번호, 사용목적, 작용원리, 원재료, 성능 등을 비교한 별지 제5호서식의 비교표</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 제2항의 자료는 다음 각 호에 해당하는 자료이여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 제2항제4호 및 제5호가목 1)·3)·5)의 자료는 다음의 어느 하나에 해당하는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 식약처장이 지정한 시험·검사기관에서 발급한 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 해당 의료기기에 대하여 경제협력개발기구(OECD) 회원국에 허가 당시 제출되어 평가된 시험성적서로서 해당 정부 또는 정부가 허가 업무를 위임한 등록기관이 제출받아 승인하였음을 확인한 자료 또는 이를 공증한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 「의료기기 제조 및 품질관리기준」(식품의약품안전처 고시) 또는 이와 동등 이상의 규격에 따른 제조사의 품질관리시스템 하에서 실시된 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 대학 또는 연구기관 등 국내·외의 전문기관에서 시험한 것으로서 해당 전문기관의 장이 발급하고 그 내용(전문기관의 시험시설 개요, 주요설비, 연구인력 구성, 시험자의 연구경력 등을 포함한다)을 검토하여 타당하다고 인정할 수 있는 시험성적서</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제2항제5호가목 2) 중 제2항제5호다목에 해당하는 시험의 자료는 다음의 어느 하나에 해당하는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 식약처장이 지정한 임상시험기관에서 시험한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 외국자료로서 그 내용을 검토하여 실시기관의 신뢰성이 인정되고 시행규칙 별표 3 의료기기 임상시험 관리기준에 의하여 실시한 시험자료 또는 이에 준하는 것으로 인정되는 시험자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 해당 의료기기에 대하여 경제협력개발기구(OECD) 회원국에 허가 당시 제출되어 평가된 임상적 성능 시험자료로서 해당 정부 또는 정부가 허가 업무를 위임한 등록기관이 제출받아 승인하였음을 확인한 자료 또는 이를 공증한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    라. 과학논문인용색인(Science Citation Index)에 등재된 전문학회지에 게재된 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 제2항제5호가목 2) 중 제2항제5호다목에 해당하는 시험을 제외한 시험의 자료는 다음의 어느 하나에 해당하는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 다음 각 단의 어느 하나에 해당하는 기관에서 별표 14 임상적 성능시험 관리기준에 의하여 실시한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    1)「감염병의 예방 및 관리에 관한 시행규칙」 제4조제9호에 해당하는 기관  </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    2)「혈액관리법」 제6조제3항에 따라 허가받은 공급혈액원</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    3) 법 제10조제3항에 따라 식약처장이 지정한 임상시험기관</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 외국자료로서 그 내용을 검토하여 실시기관의 신뢰성이 인정되고 별표 14 임상적 성능시험 관리기준에 의하여 실시한 시험자료 또는 이에 준하는 것으로 인정되는 시험자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    다. 제2호 각 목의 어느 하나에 해당하는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 제2호 가목 및 나목의 자료는 제29조제12호 나목에 관한 사항을 포함하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제34조(체외진단용 의료기기의 심사자료 면제 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 제23조제2항에도 불구하고 2등급 체외진단용 의료기기의 경우에는 별지 제5호서식의 체외진단용 의료기기의 본질적 동등품목 비교표를 사용하여 별표 6의 동등제품 판단기준에 따라 새로운제품, 개량제품, 동등제품으로 구분하여 심사한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 제1항에 따라 새로운제품, 개량제품, 동등제품으로 구분된 2등급 체외진단용 의료기기의 경우에는 별표 8에 따라 제33조제2항제1호가목·나목 및 제3호·제5호의 규정에 따른 자료의 제출을 면제할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 3·4등급 체외진단용 의료기기의 경우에는 별표 8의2에 따라 다음 각 호에 해당하는 자료의 제출을 면제할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 3등급 체외진단용 의료기기의 경우 제33조제2항제1호가목·나목 및 제5호가목 2)의 규정에 따른 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 4등급 체외진단용 의료기기 경우에는 제33조제2항제1호가목·나목의 규정에 따른 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제6장 희소의료기기의 지정 등</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제35조(희소의료기기 지정 신청)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 희소의료기기로 지정받으려는 자는 별지 제6호서식에 따른 희소의료기기 지정신청서에 다음 각 호의 서류를 첨부하여 식약처장에게 제출하여야 한다. 다만, 외국에서 희소의료기기로 지정되었음을 증명할 수 있는 입증자료 등이 있는 경우에는 제3호의 자료를 제출하지 아니할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 제36조의 희소의료기기 지정기준에 해당함을 증명하는 자료. 다만 제36조제3호 후단의 경우 용도상 특별한 효용가치를 갖는다는 것을 증명하는 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제54조제1항에 따른 해당 제품의 사용목적, 모양 및 구조, 원재료, 성능, 사용방법, 작용원리 및 규격 등에 대한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 대한의사협회장, 대한치과의사협회장, 대한한의사협회장, 대한약사회장, 대한병원협회장 또는 해당질환 관련 전문 학회장의 별지 제7호서식에 따른 희소의료기기 지정추천서. 이 경우 추천경위 및 사유, 대체치료법 또는 대체의료기기 등에 대한 의학적 견해 및 그 근거, 대상 질환에 대한 통계자료(인구대비 발생비율 등) 및 그 근거, 기타 참고의견 등의 자료를 포함하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 제1항에도 불구하고 의료기기 제조(수입) 허가 신청과 동시에 희소의료기기로 지정받으려는 자는 허가신청서 비고란에 "희소의료기기"라는 표기를 하여 신청한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제36조(희소의료기기 지정기준)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 희소의료기기의 지정기준은 다음 각 호와 같다</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 국내 환자 수(유병인구)가 20,000명 이하인 희귀질환의 치료 또는 진단 목적으로 사용될 것</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 임상적 유효성을 입증하기 위한 필요 환자 증례수가 충분하지 아니할 것</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 국내에 그 질환에 대한 대체 의약품 또는 의료기기가 없는 등 적절한 치료방법 또는 진단방법이 없거나 기타 용도상 특별한 효용가치를 갖는다고 식약처장이 인정할 것</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 개발 단계에 있는 의료기기의 경우에도 제1항에 따라 희소의료기기로 지정할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제37조(희소의료기기 지정에 따른 준수사항)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 희소의료기기 지정을 받은 의료기기 제조업자 또는 수입업자는 다음 각 호의 사항을 준수하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 법 제8조에 따라 재심사 명령을 받은 경우 「의료기기 재심사에 관한 규정」(식품의약품안전처 고시)을 따를 것</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 의료기기 첨부문서에 ‘동 제품은 희소의료기기로서 임상적 유효성이 검증 중에 있다’는 사실을 표기할 것</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제38조(희소의료기기 지정 결과 통보)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 식약처장은 제36조에 따라 심사한 결과 희소의료기기 지정기준에 적합하다고 인정되는 경우에는 대상 희귀질환과 함께 희소의료기기로 지정하고 그 결과를 신청인에게 통보하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제39조(희소의료기기 지정취소)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 식약처장은 희소의료기기로 지정된 의료기기가 다음 각 호의 어느 하나에 해당하는 경우에는 희소의료기기 지정을 취소할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 제36조에 따른 희소의료기기 지정기준에 적합하지 않은 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제38조에 따라 통보한 대상 희귀질환 이외의 목적으로 판매·임대한 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 기타 중대한 사유로 국민의 건강을 해할 우려가 있는 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제7장 의료기기 사전검토 운영</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제40조(사전검토 신청)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 사전검토를 요청하려는 자(이하 "사전검토 신청자"라 한다)는 시행규칙 제25조제2항에 따른 별지 제27호서식의 의료기기 사전검토 신청서를 작성하여 제출하여야 한다. 이 경우 첨부하는 서류 등은 전자적 기록매체(CD·디스켓 등)와 함께 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 외국의 자료는 주요사항을 발췌한 한글 요약문 및 원문을 첨부하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 사전검토 신청자는 사전검토신청서에 사전검토 신청내용과 관련된 의료기기의 개발배경 및 단계, 희망하는 회의일자, 사전검토회의에 참석을 희망하는 부서 등을 기재하고, 사전검토 내용을 구체적이고 명확하게 작성하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ④ 사전검토 신청자는 필요한 경우 사전검토 신청 전·후 해당 부서와 협의하여 사전검토가 원활히 진행될 수 있도록 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제41조(사전검토 대상 심사)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 식약처장은 사전검토 신청자가 제출한 신청서 및 요약자료에 근거하여 신청서의 신청내용이 사전검토의 대상 및 범위에 해당 되는지를 심사한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제42조(사전검토 대상 등 통보)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 식약처장은 사전검토 신청을 받은 날로부터 15일 이내에 사전검토 신청자에게 사전검토의 대상, 사전검토 회의일자, 회의자료 제출기한 등을 통보하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제43조(회의자료 제출)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 사전검토 신청자는 회의일자 30일 전까지 회의자료를 제출하여야 하며, 회의자료에는 의료기기의 기본정보(명칭, 구조, 원리, 성능, 사용목적 및 사용방법), 질문목록 및 상세질문내용, 관련 문헌 자료 등 사전검토를 위해 필요하다고 판단되는 정보자료를 포함하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 식약처장은 사전검토 신청자가 제출한 회의자료를 검토하여, 필요한 경우 추가적인 자료의 제출을 요청할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제44조(전문가 자문)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 식약처장은 사전검토 신청자가 신청한 사전검토목록 및 상세질문 내용에 대하여 외부전문가의 자문이 필요하다고 판단되는 경우 외부전문가의 의견을 들을 수 있다. </CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 외부전문가는 사전검토 회의에 참석하여 의견을 진술하거나 서면으로 의견을 제출할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 식약처장은 외부전문가의 자문을 받은 경우에는 그 필요성 및 자문결과를 이력관리에 포함하여 기록·관리한다. </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제45조(사전검토회의 실시)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 사전검토회의는 담당부서 및 사전검토 신청자의 참여 등 공개진행을 원칙으로 한다. 다만, 회의내용이 공개될 경우 기업의 이익 침해 등을 이유로 사전검토 신청자의 요청이 있는 경우에는 사전검토회의를 비공개로 진행할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 사전검토회의결과는 공개하여야 한다. 다만, 사전검토 신청자가 제출자료의 보호 등을 이유로 비공개를 요청하는 경우에는 이를 공개하지 아니할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 외부전문가는 사전검토 회의 시 지득한 비밀, 심의자료 등을 누설하거나 무단 발표 등을 하여서는 아니 된다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ④ 식약처장은 사전검토회의에 대한 회의록을 작성하고 회의록 초안을 사전검토 신청자에게 발송하여 이견의 유무를 확인하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ⑤ 식약처장은 사전검토 신청자가 회의록에 이견이 있음을 알려오는 경우에는 사전검토회의 참여자 및 사전검토 신청자와 협의를 거쳐 회의록을 수정할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제46조(사전검토 결과통보)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 식약처장은 사전검토를 완료하는 경우 시행규칙 제25조제3항에 따른 별지 제28호서식의 의료기기 사전검토 결과통지서를 사전검토가 완료된 날로부터 10일 이내에 사전검토 신청자에게 통보하여야 한다. </CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 식약처장은 사전검토 실시결과 등에 대하여 별지 제8호서식의 이력관리대장을 작성하고 사전검토 이력관리를 하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제47조(사전검토 결과의 변경)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 식약처장은 제46조에도 불구하고 다음 각 호의 어느 하나에 해당하는 경우에는 이미 통보된 사전검토결과를 변경할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 사전검토 신청자가 변경사항에 동의한 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 사전검토결과가 통보된 이후에 사전검토결과에 중대한 영향을 미치는 안전성 및 유효성과 관련된 자료, 사실 등이 확인 된 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 식약처장은 사전검토결과를 변경할 때에는 변경의 사유 및 내용을 명시하여 사전검토 신청자에게 지체 없이 서면으로 통보하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제48조(이의신청)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 사전검토 신청자는 제46조 및 제47조에 따라 통보받은 사전검토결과 또는 사전검토결과의 변경에 이의가 있는 경우에는 통보받은 날로부터 30일 이내에 별지 제9호서식의 이의신청서에 이의신청의 취지 및 이유를 명시하여 이의신청을 할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 식약처장은 제1항에 따라 이의신청을 받은 경우에는 이의신청을 받은 날로부터 20일 이내에 사전검토 신청자에게 검토결과를 서면으로 알려야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 식약처장은 사전검토 신청자의 이의신청에 관한 사항에 대하여 필요한 경우 의료기기위원회의 자문을 받을 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제8장 단계별 심사 운영</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제49조(심사 대상)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 식약처장은 제조허가를 받고자 개발 중인 제품으로서 임상시험이 필요한 다음 각 호의 어느 하나에 해당하는 의료기기에 대하여 개발 단계별로 심사할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 첨단의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 신개발 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 기타 식약처장이 필요하다고 인정하는 의료기기</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제50조(심사 신청)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 단계별 심사를 받고자 하는 자는 별지 제15호 서식의 의료기기 단계별 심사 신청서(전자문서로 된 신청서를 포함한다)에 별표 15에 따른 자료(전자문서를 포함한다)를 첨부하여 식약처장에게 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제51조(심사자료의 종류 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 단계별 심사를 위하여 제출하여야 하는 자료는 다음 각 호와 같다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 제품설계 및 개발 검토단계 자료(1단계에 한함)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 안전성 및 성능 검토단계 자료(2단계에 한함)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 임상시험계획서 검토단계 자료(3단계에 한함)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 기술문서 및 임상자료 검토단계 자료(4단계에 한함)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 제1항에 따라 단계별로 제출하는 자료의 범위는 별표 15와 같다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ③ 신청인은 해당 단계의 제출자료가 마련되는 시점에 한 개 이상의 자료를 제출할 수 있다. 이 경우 식약처장은 제출자료별로 보완을 요청할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제52조(심사 절차)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 단계별로 제출자료를 심사하여 적합한 경우 이후 단계로 심사를 진행하고, 4단계 완료시 심사가 종료된다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 식약처장은 단계별로 제출자료를 검토하여 필요한 경우 신청인에게 제품상세 개요 및 단계별 제출자료 등에 대한 검토회의를 요청할 수 있다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제53조(심사결과 통보 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 식약처장은 단계별로 제출된 자료를 검토하여 각 자료 제출일로부터 30일 내에 검토 결과를 통보하여야 한다. 다만, 외부 전문가 자문, 회의 또는 추가 제출자료 검토 등으로 인하여 필요한 경우 검토 결과의 통보 기한을 연장할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 식약처장은 4단계의 단계별 심사를 완료하면 시행규칙 별지 제9호서식의 의료기기 기술문서 등의 심사결과통지서를 신청인에게 통보하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제54조(심사결과의 변경)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 제53조에도 불구하고 식약처장은 다음 각 호의 어느 하나에 해당하는 경우에는 이미 통보된 심사결과를 변경할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 제품 개발도중 안전성 및 유효성에 영향을 미치는 변경사항이 발생한 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 안전성 정보가 새롭게 제시되어 제품의 안전성 및 유효성의 확보가 필요한 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 신청인이 요청하여 변경의 필요성이 인정되는 경우</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 신청인은 제1항에 따라 심사 자료를 변경할 때에는 이전 단계의 변경 자료를 다시 제출하고 식약처장은 심사한 결과에 대해 통보하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제9장 판매업 신고가 면제되는 의료기기 지정</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제55조(의료기기 판매업 신고가 면제되는 의료기기)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 시행규칙 제38조제3호에 따라 의료기기 판매업 신고를 하지 아니하고 판매할 수 있는 의료기기는 다음 각 호와 같다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 전자체온계</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 귀 적외선 체온계</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 피부 적외선 체온계</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  4. 자동 전자혈압계</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  5. 자가진단용 모바일 의료용 애플리케이션(이하 "앱"이라 한다) 및 이를 탑재한 제품(휴대전화, 태블릿 PC 등)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  6. 개인용체외진단검사시약 I(임신진단용에 한함)</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제10장 전시 목적 의료기기의 승인</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제56조(승인신청)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 시행규칙 제46조에 따라 전시할 목적의 의료기기(이하 "전시용 의료기기"라 한다)를 진열하고자 하는 자는  다음 각 호의 서류를 첨부하여 관할 지방청장에게 승인을 받아야 한다. 다만, 전시회를 주관 또는 주최하는 의료기기 관련 단체 등에서 그 수요를 파악하여 일괄해서 승인을 신청할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 전시목적, 전시장소, 전시기간 등을 포함한 전시계획에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제품설명 및 홍보자료로 다음 각 목에 해당하는 자료 </CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 사용목적에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 제품의 모양 및 구조, 성능, 특성, 작용원리, 사용방법에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 전시용 의료기기임을 표시하기 위한 기재 및 부착방법에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제57조(승인기준)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 지방청장은 제50조에 따른 전시용 의료기기 승인신청을 받은 경우에는 다음 각 호의 기준에 따라 적합 여부를 검토하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 국가 또는 지방자치단체, 의료기기 관련 단체가 주관 또는 주최하는 박람회, 전람회, 전시회 및 그 밖에 이에 준하는 전시회 이어야 할 것</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 제품설명 및 홍보자료는 법 제24조제1항제1호 및 제3호, 제2항제1호부터 제4호까지의 규정에 따라 금지하고 있는 내용을 포함하지 아니할 것</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. "전시용 의료기기" 또는 "허가받지 않은 의료기기"라는 표시가 누구나 쉽게 보거나 알 수 있도록 부착되어야 할 것</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제58조(승인통보)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 지방청장은 신청일로부터 10일 이내 승인여부를 문서로 통보하여야 하며, 승인사항을 전시회 등이 개최되는 관할 지방청장에게 통보하여야 한다. </CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 승인통보 시에는 다음 각 호의 사항을 포함하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 전시회 등의 명칭 및 장소</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 전시 승인 기간 및 해당 제품</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 승인된 전시용 의료기기에 대한 다음 각 목의 허용범위에 대한 사항</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    가. 사용방법이나 사용목적 등을 설명하기 위한 단순 구동</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="38" Style="0"><TEXT CharShape="5"><CHAR>    나. 전시장내에서 승인받은 제품의 설명·홍보자료의 부착 및 비치</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제11장 의료기기의 성능 개선 허용 범위</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제59조(성능개선 허용 대상)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 소프트웨어의 성능을 개선하려는 의료기기 또는 소프트웨어의 성능 개선과 관련한 하드웨어를 변경하려는 의료기기는 법 제26조제4항 단서 및 시행규칙 제47조에 따라 제조업자와 수입업자가 변경 허가·인증을 받거나 변경 신고한 범위 내에서 성능 개선 할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제12장 의료기기 해당 여부 검토</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제60조(의료기기 해당 여부 검토 신청 등)</CHAR></TEXT><TEXT CharShape="5"><CHAR> ① 의료기기 허가·인증·신고·심사 등을 위하여 어떤 제품이 법 제2조제1항에 따른 의료기기에 해당되는지 검토해 줄 것을 의뢰하고자 하는 자는 다음 각 호의 자료를 갖추어 식약처장에게 제출하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  1. 그 제품의 사용 목적에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  2. 그 제품의 모양 및 구조, 원재료, 성능, 사용방법 등에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="37" Style="0"><TEXT CharShape="5"><CHAR>  3. 기타 그 제품에 대한 작용원리 및 규격 등에 관한 자료</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="36" Style="0"><TEXT CharShape="5"><CHAR>  ② 식약처장은 제1항의 검토 의뢰가 있는 경우에는 그 제품이 법 제2조제1항에 부합하는지, 「의료기기 품목 및 품목별 등급에 관한 규정」에 따른 개별 품목에 해당하는지, 등급 분류와 지정을 할 수 있는지 등을 검토하고, 그 결과를 10일 이내에 신청인에게 통보하여야 한다.</CHAR></TEXT><TEXT CharShape="7" /></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>        제13장 보 칙</CHAR></TEXT></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제61조(자문)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 식약처장은 허가·인증·신고·심사를 위하여 필요한 경우에는 의료기기위원회 및 식품의약품안전평가원의 자문을 받을 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제62조(제출자료의 신뢰성 확인)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 식약처장은 제29조 또는 제33조에 따라 제출한 시험성적서 또는 자료에 대하여 신뢰성 확인이 필요한 경우에는 현장 확인을 실시할 수 있다.</CHAR></TEXT><TEXT CharShape="7"/></P><P ParaShape="23"><TEXT CharShape="6"><CHAR>제63조(규제의 재검토)</CHAR></TEXT><TEXT CharShape="5"><CHAR> 「행정규제기본법」제8조 및 「훈령·예규 등의 발령 및 관리에 관한 규정」(대통령훈령 제248호)에 따라 2014년 1월 1일을 기준으로 매 3년이 되는 시점(매 3년째의 12월 31일까지를 말한다)마다 그 타당성을 검토하여 개선 등의 조치를 하여야 한다.</CHAR></TEXT><TEXT CharShape="7"/></P>
<P ParaShape="2"><TEXT CharShape="5"/></P><P ParaShape="2"><TEXT CharShape="6"><CHAR>&nbsp;&nbsp;&nbsp;&nbsp;부칙&nbsp;</CHAR></TEXT><TEXT CharShape="23"><CHAR>&nbsp;&lt;제2017-58호, 2017.7.10&gt; &nbsp;</CHAR></TEXT></P><P ParaShape="23" Style="0"><TEXT CharShape="6"><CHAR>제1조</CHAR></TEXT><TEXT CharShape="5"><CHAR>(시행일) 이 고시는 고시한 날부터 시행한다.</CHAR></TEXT></P><P ParaShape="23" Style="0"><TEXT CharShape="6"><CHAR>제2조</CHAR></TEXT><TEXT CharShape="5"><CHAR>(허가·인증·신고의 신청 등에 관한 경과조치) 이 고시 시행일 당시 종전의 규정에 따라 의료기기 제조(수입) 허가신청서, 의료기기 허가사항 변경허가신청서, 의료기기 제조(수입) 인증신청서, 의료기기 인증사항 변경인증신청서, 의료기기 제조(수입) 신고서, 의료기기 기술문서 등 심사의뢰서를 접수한 경우에는 개정 규정에도 불구하고 종전의 규정에 따른다.</CHAR></TEXT></P>


</SECTION>
</BODY>
<TAIL><BINDATASTORAGE><BINDATA Size="84377" Encoding="Base64" Id="1">iVBORw0KGgoAAAANSUhEUgAAAT0AAADECAYAAADtcfb4AAAACXBIWXMAAAsTAAALEwEAmpwYAAAK
[encoded data, 13984 characters]
cd5u+Ztqm21hi4FITxAEoYCRysmCIIj0BEEQRHqCIAgFwP8NAEQiqcYk0yx6AAAAAElFTkSuQmCC</BINDATA>
<BINDATA Size="5750" Encoding="Base64" Id="2">Qk1mEAAAAAAAADYAAAAoAAAAJQAAACUAAAABABgAAAAAADAQAAAAAAAAAAAAAAAAAAAAAAAA
[encoded data, 5472 characters]
////////////////////////////////////////////////////AA==</BINDATA>
<BINDATA Size="84377" Encoding="Base64" Id="3">iVBORw0KGgoAAAANSUhEUgAAAT0AAADECAYAAADtcfb4AAAACXBIWXMAAAsTAAALEwEAmpwYAAAK
[encoded data, 13984 characters]
cd5u+Ztqm21hi4FITxAEoYCRysmCIIj0BEEQRHqCIAgFwP8NAEQiqcYk0yx6AAAAAElFTkSuQmCC</BINDATA>
<BINDATA Size="252903" Encoding="Base64" Id="4">[encoded data, 9304 characters]</BINDATA><BINDATA Size="252903" Encoding="Base64" Id="5">[encoded data, 6468 characters]</BINDATA><BINDATA Size="252903" Encoding="Base64" Id="6">[encoded data, 6468 characters]</BINDATA><BINDATA Size="252903" Encoding="Base64" Id="7">[encoded data, 5656 characters]</BINDATA><BINDATA Size="252903" Encoding="Base64" Id="8">[encoded data, 8260 characters]</BINDATA>
<BINDATA Size="252903" Encoding="Base64" Id="9">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</BINDATA>
</BINDATASTORAGE>
</TAIL>
</HWPML>
